# Message Decoder for Solo2, Alto and Deep Solo Floats; v260708
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11011     8  258 000000000000009  3874     0   6 2020/04/12, 22:24:14,  41.5334  -70.6398   4      0 6,14,14
SBD 11011     8  340 000000000000009  3875     0   5 2020/04/12, 22:24:48,  41.5413  -70.6851   3      0 13
SBD 11011     8  333 000000000000009  3876     0   4 2020/04/12, 22:24:59,  41.5413  -70.6851   3      0 8,15,13
SBD 11011     8  335 000000000000009  3877     0   3 2020/04/12, 22:25:12,  41.5413  -70.6851   3      0 8
SBD 11011     8  279 000000000000009  3878     0   2 2020/04/12, 22:25:32,  41.5413  -70.6851   3      0 10,11,0,1,2,3
SBD 11011     8  305 000000000000009  3879     0   1 2020/04/12, 22:25:52,  41.5413  -70.6851   3      0 10,11,7,5,14
SBD 11011     8  333 000000000000009  3880     0   0 2020/04/12, 22:26:06,  41.5683  -70.6851   3      0 0,4,14,14,9,9
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      25         2099           84        2183
# Sensor_ID bytes     % packets Description
          0    48  2.29       2         GPS
          1    33  1.57       1    Pressure
          2    37  1.76       1 Temperature
          3    33  1.57       1    Salinity
          4   133  6.34       1        Fall
          5    58  2.76       1        Rise
          6   158  7.53       1        Pump
          7    72  3.43       1 Drift_times
          8   502 23.92       2     Surface
          9   120  5.72       2  HR/Drift_P
         10   142  6.77       2  HR/Drift_T
         11   142  6.77       2  HR/Drift_S
         13   404 19.25       2  Parameters
         14   151  7.19       5 Engineering
         15    66  3.14       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G    8 1 2020/04/10 19:26 +41.53428  -70.64684    0  7 28 33 44  1.1     0
G    8 2 2020/04/12 22:13 +41.53433  -70.64679   50  5 31 35 41  1.8     0
# Profile Data
#      C Npts comp n_var sub_cycle
D0     8   10    1     3         2
#      Pres      Temp    Salinity
      1.480    17.7960    30.2100
      2.080    17.1090    30.1340
      3.000    16.6680    30.1030
      4.000    16.3880    30.0880
      5.040    16.1230    30.0790
      6.080    15.8990    30.0680
      7.080    15.7360    30.0580
      8.000    15.2830    30.0330
      9.160    13.2980    30.0320
      9.600    12.4930    30.0210
# High Resolution Data
#     C   Npts comp
H0     8   47     1
#      Pres      Temp    Salinity
      1.440    17.9030    30.2120
      1.480    17.6890    30.2090
      1.640    17.5210    30.1750
      1.640    17.4030    30.1580
      1.760    17.3210    30.1410
      1.880    17.2440    30.1360
      1.920    17.1800    30.1310
      2.080    17.1200    30.1290
      2.080    17.0530    30.1340
      2.200    16.9900    30.1300
      2.280    16.9250    30.1320
      2.320    16.8770    30.1250
      2.480    16.8490    30.1080
      2.480    16.8280    30.1050
      2.600    16.8080    30.1050
      2.680    16.7810    30.1080
      2.720    16.7460    30.1110
      2.880    16.7140    30.1100
      2.880    16.6830    30.1010
      3.000    16.6590    30.1010
      3.120    16.6400    30.0960
      3.120    16.6180    30.1000
      3.280    16.5920    30.1020
      3.360    16.5640    30.0960
      3.400    16.5390    30.1010
      3.520    16.5100    30.1010
      3.520    16.4820    30.0990
      3.680    16.4620    30.0920
      3.680    16.4380    30.0940
      3.800    16.4210    30.0880
      3.880    16.4030    30.0890
      3.920    16.3910    30.0850
      4.080    16.3790    30.0820
      4.080    16.3640    30.0870
      4.200    16.3510    30.0830
      4.240    16.3320    30.0870
      4.280    16.3150    30.0840
      4.440    16.3000    30.0800
      4.440    16.2820    30.0840
      4.560    16.2680    30.0810
      4.600    16.2420    30.0880
      4.640    16.2180    30.0810
      4.760    16.1970    30.0850
      4.760    16.1800    30.0780
      4.920    16.1610    30.0830
      4.920    16.1350    30.0840
      4.960    16.1140    30.0830
# Drift Data
#      C Npts comp n_var
I0     8   96    1     4
#      Pres       Temp   Salinity     Time
      9.280    12.7020    30.0260        6
      9.320    12.6990    30.0250     1812
      9.320    12.6970    30.0250     3618
      9.320    12.6950    30.0250     5423
      9.320    12.6910    30.0250     7229
      9.360    12.6880    30.0250     9035
      9.400    12.6860    30.0250    10841
      9.360    12.6850    30.0240    12647
      9.400    12.6810    30.0250    14452
      9.360    12.6790    30.0240    16258
      9.400    12.6740    30.0250    18064
      9.400    12.6730    30.0240    19870
      9.400    12.6700    30.0250    21676
      9.400    12.6670    30.0250    23481
      9.400    12.6630    30.0250    25287
      9.400    12.6610    30.0240    27093
      9.400    12.6580    30.0240    28899
      9.440    12.6540    30.0250    30705
      9.400    12.6510    30.0250    32511
      9.400    12.6470    30.0250    34316
      9.400    12.6450    30.0240    36122
      9.400    12.6410    30.0250    37928
      9.400    12.6390    30.0240    39734
      9.440    12.6340    30.0240    41540
      9.440    12.6310    30.0250    43345
      9.440    12.6280    30.0240    45151
      9.440    12.6240    30.0240    46957
      9.440    12.6200    30.0250    48763
      9.440    12.6170    30.0250    50569
      9.440    12.6130    30.0250    52374
      9.440    12.6110    30.0240    54180
      9.440    12.6090    30.0240    55986
      9.440    12.6050    30.0250    57792
      9.440    12.6020    30.0250    59598
      9.440    12.5990    30.0250    61403
      9.440    12.5970    30.0240    63209
      9.480    12.5930    30.0250    65015
      9.440    12.5910    30.0240    66821
      9.440    12.5880    30.0250    68627
      9.440    12.5870    30.0240    70432
      9.440    12.5820    30.0250    72238
      9.480    12.5800    30.0250    74044
      9.440    12.5780    30.0240    75850
      9.440    12.5760    30.0250    77656
      9.440    12.5740    30.0250    79461
      9.480    12.5710    30.0250    81267
      9.440    12.5700    30.0240    83073
      9.440    12.5680    30.0230    84879
      9.480    12.5650    30.0250    86685
      9.480    12.5630    30.0240    88490
      9.480    12.5610    30.0250    90296
      9.480    12.5620    30.0240    92102
      9.520    12.5560    30.0250    93908
      9.520    12.5560    30.0240    95714
      9.520    12.5530    30.0240    97519
      9.520    12.5510    30.0240    99325
      9.520    12.5480    30.0250   101131
      9.520    12.5470    30.0240   102937
      9.520    12.5450    30.0240   104743
      9.520    12.5450    30.0240   106548
      9.560    12.5400    30.0250   108354
      9.560    12.5380    30.0240   110160
      9.520    12.5370    30.0240   111966
      9.560    12.5340    30.0250   113772
      9.560    12.5310    30.0250   115577
      9.560    12.5270    30.0250   117383
      9.560    12.5260    30.0250   119189
      9.560    12.5250    30.0240   120995
      9.600    12.5210    30.0250   122801
      9.600    12.5180    30.0250   124606
      9.560    12.5170    30.0240   126412
      9.560    12.5120    30.0250   128218
      9.560    12.5100    30.0250   130024
      9.600    12.5070    30.0250   131830
      9.600    12.5040    30.0240   133635
      9.600    12.5030    30.0240   135441
      9.600    12.5000    30.0240   137247
      9.600    12.4970    30.0250   139053
      9.600    12.4950    30.0250   140858
      9.600    12.4930    30.0250   142664
      9.600    12.4910    30.0250   144470
      9.600    12.4930    30.0240   146276
      9.600    12.4900    30.0250   148082
      9.600    12.4860    30.0260   149887
      9.600    12.4840    30.0250   151693
      9.600    12.4830    30.0260   153499
      9.600    12.4810    30.0250   155305
      9.600    12.4780    30.0260   157111
      9.600    12.4780    30.0250   158916
      9.600    12.4750    30.0260   160722
      9.600    12.4750    30.0250   162528
      9.600    12.4750    30.0250   164334
      9.600    12.4730    30.0260   166140
      9.600    12.4730    30.0250   167945
      9.600    12.4740    30.0250   169751
      9.600    12.4740    30.0250   171557
# Surface Pressure Time Series
#    C npts n_var               Start
S    8    54     6 2020/04/12 22:13:51
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
    73      0.240     21.960      0.000      14 2020/04/12 22:13:51
    86      0.250     22.033      0.000      14 2020/04/12 22:14:04
    97      0.250     22.137      0.000      14 2020/04/12 22:14:15
   108      0.240     22.263      0.000      14 2020/04/12 22:14:26
   119      0.230     22.378      0.000      14 2020/04/12 22:14:37
   130      0.230     22.466      0.000      14 2020/04/12 22:14:48
   141      0.240     22.502      0.000      14 2020/04/12 22:14:59
   153      0.250     22.562      0.000      14 2020/04/12 22:15:11
   164      0.250     22.577      0.000      14 2020/04/12 22:15:22
   175      0.240     22.621      0.000      14 2020/04/12 22:15:33
   186      0.230     22.648      0.000      14 2020/04/12 22:15:44
   197      0.250     22.637      0.000      14 2020/04/12 22:15:55
   208      0.230     22.673      0.000      14 2020/04/12 22:16:06
   219      0.240     22.665      0.000      14 2020/04/12 22:16:17
   230      0.240     22.630      0.000      14 2020/04/12 22:16:28
   241      0.230     22.644      0.000      14 2020/04/12 22:16:39
   252      0.250     22.674      0.000      14 2020/04/12 22:16:50
   264      0.230     22.685      0.000      14 2020/04/12 22:17:02
   275      0.230     22.637      0.000      14 2020/04/12 22:17:13
   286      0.240     22.635      0.000      14 2020/04/12 22:17:24
   297      0.240     22.669      0.000      14 2020/04/12 22:17:35
   308      0.230     22.665      0.000      14 2020/04/12 22:17:46
   319      0.250     22.668      0.000      14 2020/04/12 22:17:57
   330      0.240     22.691      0.000      14 2020/04/12 22:18:08
   341      0.230     22.731      0.000      14 2020/04/12 22:18:19
   352      0.250     22.723      0.000      14 2020/04/12 22:18:30
   363      0.230     22.730      0.000      14 2020/04/12 22:18:41
   374      0.250     22.756      0.000      14 2020/04/12 22:18:52
   386      0.230     22.739      0.000      14 2020/04/12 22:19:04
   397      0.250     22.695      0.000      14 2020/04/12 22:19:15
   408      0.240     22.658      0.000      14 2020/04/12 22:19:26
   419      0.240     22.657      0.000      14 2020/04/12 22:19:37
   430      0.250     22.689      0.000      14 2020/04/12 22:19:48
   441      0.230     22.718      0.000      14 2020/04/12 22:19:59
   452      0.230     22.726      0.000      14 2020/04/12 22:20:10
   463      0.230     22.722      0.000      14 2020/04/12 22:20:21
   474      0.220     22.720      0.000      14 2020/04/12 22:20:32
   485      0.220     22.703      0.000      14 2020/04/12 22:20:43
   496      0.250     22.706      0.000      14 2020/04/12 22:20:54
   508      0.250     22.771      0.000      14 2020/04/12 22:21:06
   519      0.230     22.826      0.000      14 2020/04/12 22:21:17
   530      0.230     22.815      0.000      14 2020/04/12 22:21:28
   541      0.240     22.813      0.000      14 2020/04/12 22:21:39
   552      0.240     22.845      0.000      14 2020/04/12 22:21:50
   563      0.230     22.851      0.000      14 2020/04/12 22:22:01
   574      0.230     22.872      0.000      14 2020/04/12 22:22:12
   585      0.220     22.881      0.000      14 2020/04/12 22:22:23
   596      0.240     22.841      0.000      14 2020/04/12 22:22:34
   607      0.240     22.845      0.000      14 2020/04/12 22:22:45
   618      0.230     22.874      0.000      14 2020/04/12 22:22:56
   629      0.240     22.893      0.000      14 2020/04/12 22:23:07
   641      0.230     22.928      0.000      14 2020/04/12 22:23:19
   652      0.240     22.913      0.000      14 2020/04/12 22:23:30
   663      0.230     22.954      0.000      14 2020/04/12 22:23:41
# Fall Time Series
#  Cycle npts     Start                Date
F00    8   25 639862007 2020/04/10 19:26:47
# time     Pres phase                Date
     0   -0.120     1 2020/04/10 19:26:47
    47    1.040    10 2020/04/10 19:27:34
    60    2.320    10 2020/04/10 19:27:47
    69    3.280    10 2020/04/10 19:27:56
    78    4.320    10 2020/04/10 19:28:05
    87    5.320    10 2020/04/10 19:28:14
    93    6.040     2 2020/04/10 19:28:20
    96    6.360    10 2020/04/10 19:28:23
   103    7.200    10 2020/04/10 19:28:30
   110    8.080    10 2020/04/10 19:28:37
   121    9.080    10 2020/04/10 19:28:48
  1022    9.280    10 2020/04/10 19:43:49
  1901    9.280     3 2020/04/10 19:58:28
  1924    9.280    10 2020/04/10 19:58:51
  2825    9.280    10 2020/04/10 20:13:52
  3727    9.280    10 2020/04/10 20:28:54
  3739    9.280     3 2020/04/10 20:29:06
  4628    9.280    10 2020/04/10 20:43:55
  5530    9.280    10 2020/04/10 20:58:57
  5578    9.280     3 2020/04/10 20:59:45
  6431    9.320    10 2020/04/10 21:13:58
  7332    9.280    10 2020/04/10 21:28:59
  8234    9.280    10 2020/04/10 21:44:01
  9135    9.280    10 2020/04/10 21:59:02
  9216    9.280    10 2020/04/10 22:00:23
# Rise Time Series
#  Cycle npts     Start                Date
R00    8   10 640044037 2020/04/12 22:00:37
# time     Pres phase                Date
     0    9.600     7 2020/04/12 22:00:37
   487    9.000    13 2020/04/12 22:08:44
   559    7.920    13 2020/04/12 22:09:56
   631    6.880    13 2020/04/12 22:11:08
   658    5.920    13 2020/04/12 22:11:35
   676    4.960    13 2020/04/12 22:11:53
   691    3.920    13 2020/04/12 22:12:08
   703    3.000    13 2020/04/12 22:12:20
   718    1.760    13 2020/04/12 22:12:35
   719    1.640     8 2020/04/12 22:12:36
# Pump Time Series
#   Cycle npts
B02     8   14
# Pressure time Voltage Current Vacuum       phase
#     dBar  sec    volt      mA Before After
     6.040    5   11.94     213    624    -9     2
     9.600   31   11.96     231    639    -9     7
     9.560   12   12.00     221    640    -9    13
     9.560   12   12.01     205    641    -9    13
     9.600   12   11.95     230    642    -9    13
     9.560   12   11.92     219    643    -9    13
     9.560   12   11.97     212    644    -9    13
     9.600   12   11.97     215    645    -9    13
     9.280   12   11.97     208    645    -9    13
     8.400   12   11.95     214    646    -9    13
     7.720   12   11.94     223    644    -9    13
     6.320   12   11.93     225    642    -9    13
     2.280    6   11.93     218    639    -9    13
    -0.080   59   12.00     178    644    -9    14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2     8  2.4    4  2000    10   600    32   600  2880      5     3    1725   30  2.9380
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17    8 2020/04/10 19:26:44
# Pcase vacuums
   6.440   6.380
# Oil valve positions
   1.767   0.855   0.602
# Seek attempts and pump time
      3      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED18    8 2020/04/10 22:00:24 9999/99/99 99:99:99
# Pcase vacuums
   6.370   6.390
# Average 
#   P      T      S
    9.480  12.574  30.025
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA18    8 2020/04/12 22:00:37 9999/99/99 99:99:99
# Pcase vacuums
   6.390   6.390
# Alto Engineering: Surface, version 20
#       C Surface Start Date
EX20    8 2020/04/12 22:12:35
# number of packets previous packets
      7              12
# CTD Status pres offset
 0x4e410000   0.000
# Pcase vacuums
   6.390   6.350
# Air Pump max current time  max pressure
       0         0     1.289
# Air Valve Positions
   1.166   0.000
# Air Valve Calibrations
   1.103   0.086   0.595
# Oil Valve Positions
   0.000   0.854
# Oil Valve Calibrations
   1.735   0.869
# Encountered Error Codes
#    Cycle Errors
EE20     8      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00    8  1      0
# MC    Pres      Time  Pcase Vacuum
   1    -0.120     87   6.240
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M    8       197    2020/04/12 22:24:59
Parameter Table Param_Table-V17.txt |
     IFV =  7.2.9.0	(IFV) |
     VER =       17	(version) |
     STY =        1	(Sensor 0=RBR,1=SBE) |
     MSN =        8	(cycle number) |
     MPS =        2	(phase) |
     MPC =        1	(count) |
  MNM[0] =        3	(count) |
  MNM[1] =        3	(count) |
  MNM[2] =        3	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =       10	(dBar) |
  DTD[1] =       10	(dBar) |
  DTD[2] =       10	(dBar) |
     MDS =     2200	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        3	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =        5	(dBar) |
     CRS =     2000	(HRD samples) |
     MTD =        2	(dBar) |
     MTT =   150000	(ms) |
     MTS =    21600	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =     1200	(V) |
     PNC =      400	(inHg) |
     PXC =     1500	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        5	(summer attempts) |
  ICA[1] =        5	(winter attempts) |
  ICB[0] =        1	(summer cycles) |
  ICB[1] =        1	(winter cycles) |
  IMT[0] =       70	(summer message threshold) |
  IMT[1] =       70	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       20	(shallow limit for test dBar) |
     ICX =       40	(deep limit for test dBar) |
     ICT =     0.00	(deg C) |
     ICS =       30	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      180	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       30	(summer minutes) |
  ICR[1] =       30	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =       45	(summer secs) |
  ICO[1] =       45	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =        7	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =        5	(dBar) |
  DSD[1] =        5	(dBar) |
  DSD[2] =        5	(dBar) |
  DST[0] =     1800	(sec) |
  DST[1] =     1800	(sec) |
  DST[2] =     1800	(sec) |
     DDF =       60	(sec) |
     DDC =        0	(sec) |
     DSI =     2000	(ms) |
     DDD =        1	(dBar) |
     DDT =      900	(sec) |
  DFT[0] =        1	(sec) |
  DFT[1] =        1	(sec) |
  DFT[2] =        2	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =     1800	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =     1800	(sec) |
     DCA =        3	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =        0	(sec) |
     DCX =     1020	(sec) |
     DCP =        2	(sec) |
     DCF =        2	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =     1800	(sec) |
     DUC =       50	(under damping factor) |
     DPT =       10	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =      900	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =        3	(dBar) |
     DDS =      120	(sec) |
     PNS =        7	(cycle) |
  PTO[0] =    10800	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   172800	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =     2048	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =  1800000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        0	(0=dis 1=en) |
  PFT[0] =        1	(sec) |
  PFT[1] =        1	(sec) |
  PFT[2] =    36000	(sec) |
  PFD[0] =      500	(dBar) |
  PFD[1] =     1000	(dBar) |
  PFD[2] =     2000	(dBar) |
  PFV[0] =        1	(sec) |
  PFV[1] =        1	(sec) |
  PFV[2] =      300	(sec) |
     PDT =     1800	(sec) |
     PSM =        1	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    21600	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     PSS =        7	(1=P,2=T,3=PT,4=S,5=PS,6=TS,7=PTS) |
     ASN =        7	(cycle number) |
     ARR =        5	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APN =       30	(sec) |
     APF =       10	(sec) |
     ASI =  3276800	(ms) |
     ASD =     2000	(dBar) |
     ADD =        2	(dBar) |
     ADT =        1	(sec) |
     ARC =      900	(?) |
     AHR =     4500	(cm/sec) |
     AHD =     1000	(dBar) |
     SFN =       10	(cycle number) |
     SDC =        7	(dBar) |
     SGT =        0	(min) |
     SET =       15	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =     3840	(sec) |
     SSA =      300	(0=dis 1=en) |
     SSI =        0	(ms) |
     SDS =    10000	(sec) |
     SAM =        0	(0=dis 1=en) |
     SAX =     2561	(.001 psi) |
     SAN =     7500	(.001 psi) |
     SAO =     6500	(sec) |
     SAP =     1080	(sec) |
     SPT =      600	(sec) |
     SSP =       60	(sec) |

