# Message Decoder for Solo2, Alto and Deep Solo Floats; v210127
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11010   298   86 000000000233020  3309     0   6 2020/08/26, 02:51:40,  74.3562 -174.4084 139      0 14
SBD 11010   298  337 000000000233020  3310     0   5 2020/08/26, 02:52:16,  74.5000 -178.3902   6      0 13,6,14
SBD 11010   298  340 000000000233020  3311     0   4 2020/08/26, 02:52:42,  74.5000 -178.3902   5      0 13
SBD 11010   298  292 000000000233020  3312     0   3 2020/08/26, 02:55:57,  74.4654 -178.3940   3      0 3,8
SBD 11010   298  288 000000000233020  3313     0   2 2020/08/26, 02:56:21,  74.5000 -178.3902   3      0 1,2
SBD 11010   298  325 000000000233020  3314     0   1 2020/08/26, 02:56:38,  74.5000 -178.3902   3      0 10,11,5,14,0,15
SBD 11010   298  300 000000000233020  3315     0   0 2020/08/26, 02:56:55,  74.4644 -178.2602  19      0 0,4,14,14,9
# Delayed Transmission - (7/7) Messages transmitted 205.1 days after surfacing on 2020/02/03 00:38:57 (refer to 1st Rise scan < 3dbar)
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      20         1884           84        1968
# Sensor_ID bytes     % packets Description
          0    48  2.55       2         GPS
          1   106  5.63       1    Pressure
          2   170  9.02       1 Temperature
          3   146  7.75       1    Salinity
          4   138  7.32       1        Fall
          5    58  3.08       1        Rise
          6   213 11.31       1        Pump
          8   134  7.11       1     Surface
          9    76  4.03       1  HR/Drift_P
         10    76  4.03       1  HR/Drift_T
         11    76  4.03       1  HR/Drift_S
         13   407 21.60       2  Parameters
         14   170  9.02       5 Engineering
         15    66  3.50       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  298 1 9999/99/99 99:99 +99.99000 +999.99000    0  0  0  0  0  0.0     0
G  298 2 9999/99/99 99:99 +99.99000 +999.99000    0  0  0  0  0  0.0     0
# Profile Data
#      C Npts comp n_var
D0   298  164    1     3
#      Pres      Temp    Salinity
      1.080    -1.7350    31.7470
      2.000    -1.7350    31.7470
      3.000    -1.7350    31.7470
      3.920    -1.7350    31.7470
      4.960    -1.7350    31.7470
      5.920    -1.7350    31.7470
      7.000    -1.7350    31.7470
      8.040    -1.7340    31.7460
      9.080    -1.7340    31.7470
     10.040    -1.7340    31.7460
     11.000    -1.7340    31.7460
     12.000    -1.7340    31.7460
     13.000    -1.7340    31.7460
     14.080    -1.7340    31.7460
     15.080    -1.7340    31.7460
     16.000    -1.7340    31.7460
     16.960    -1.7340    31.7470
     17.960    -1.7340    31.7470
     18.960    -1.7340    31.7470
     20.040    -1.7340    31.7470
     21.080    -1.7340    31.7470
     22.040    -1.7340    31.7470
     22.960    -1.7340    31.7470
     24.000    -1.7340    31.7470
     25.080    -1.7340    31.7470
     26.000    -1.7340    31.7470
     26.920    -1.7340    31.7470
     27.960    -1.7340    31.7470
     29.040    -1.7340    31.7470
     30.080    -1.7340    31.7470
     31.080    -1.7340    31.7470
     32.120    -1.7340    31.7470
     33.080    -1.7340    31.7480
     33.960    -1.7340    31.7480
     34.960    -1.7340    31.7480
     36.040    -1.7340    31.7490
     37.000    -1.7340    31.7490
     37.960    -1.7340    31.7490
     39.000    -1.7340    31.7500
     40.080    -1.7340    31.7500
     41.040    -1.7340    31.7500
     41.920    -1.7330    31.7500
     43.000    -1.7330    31.7510
     44.040    -1.7330    31.7510
     45.000    -1.7330    31.7520
     45.960    -1.7330    31.7520
     46.880    -1.7330    31.7520
     47.920    -1.7320    31.7530
     49.000    -1.7300    31.7560
     50.040    -1.7300    31.7570
     50.960    -1.7300    31.7590
     51.920    -1.7230    31.7640
     53.000    -1.6810    31.7810
     54.040    -1.6270    31.8000
     54.960    -1.5500    31.8250
     55.960    -1.5030    31.8390
     57.000    -1.3710    31.8820
     58.040    -1.2680    31.9300
     59.040    -1.1350    31.9930
     60.000    -0.9800    32.0820
     60.960    -0.8640    32.1610
     61.960    -0.8150    32.2090
     63.000    -0.8200    32.2400
     64.000    -0.8370    32.2610
     64.960    -0.8100    32.2830
     65.920    -0.8160    32.3090
     66.880    -0.7400    32.3160
     67.880    -0.7020    32.3410
     68.960    -0.6250    32.3630
     70.000    -0.6080    32.3950
     71.000    -0.7770    32.4050
     72.040    -0.8430    32.4340
     73.080    -0.8750    32.4700
     74.000    -0.8710    32.4850
     75.000    -0.8580    32.5050
     76.040    -0.7980    32.5240
     77.000    -0.7400    32.5510
     78.000    -0.7220    32.5750
     78.960    -0.8110    32.6060
     79.960    -0.7270    32.6140
     81.000    -0.6180    32.6420
     81.960    -0.4970    32.6630
     82.960    -0.4410    32.6740
     83.920    -0.4140    32.6840
     84.960    -0.3850    32.6950
     85.960    -0.3950    32.7080
     86.880    -0.4150    32.7120
     87.920    -0.4590    32.7310
     88.920    -0.5710    32.7410
     89.920    -0.7080    32.7500
     90.960    -0.8140    32.7630
     92.000    -0.8130    32.7830
     93.080    -0.8010    32.8160
     94.040    -0.8060    32.8440
     95.000    -0.8680    32.8860
     95.960    -0.8850    32.9450
     96.960    -0.8680    33.0020
     98.000    -0.8650    33.0930
     99.000    -0.8630    33.1440
     99.960    -0.9050    33.1940
    101.040    -0.9240    33.2500
    102.080    -0.9180    33.2870
    103.080    -0.9090    33.3240
    104.080    -0.8950    33.3660
    105.080    -0.8780    33.4010
    106.040    -0.8100    33.4840
    107.040    -0.7660    33.5980
    108.000    -0.7400    33.6780
    108.920    -0.7040    33.7710
    109.960    -0.6700    33.8590
    110.960    -0.6110    33.9470
    111.960    -0.5610    34.0080
    112.960    -0.5170    34.0680
    114.000    -0.4950    34.0910
    115.000    -0.4290    34.1660
    115.920    -0.3990    34.2040
    116.960    -0.3710    34.2330
    118.040    -0.3360    34.2640
    119.000    -0.3110    34.2920
    119.960    -0.2980    34.3120
    120.960    -0.2890    34.3340
    121.920    -0.2930    34.3590
    122.920    -0.3090    34.3800
    124.000    -0.3220    34.4020
    125.040    -0.3240    34.4210
    126.000    -0.3210    34.4270
    127.000    -0.3190    34.4280
    128.040    -0.3360    34.4500
    129.080    -0.3420    34.4560
    130.040    -0.3340    34.4610
    131.040    -0.3180    34.4670
    132.000    -0.3080    34.4720
    133.000    -0.2990    34.4750
    133.960    -0.2970    34.4750
    135.000    -0.2860    34.4770
    136.040    -0.2800    34.4800
    137.000    -0.2790    34.4820
    137.960    -0.2780    34.4830
    138.960    -0.2770    34.4850
    140.040    -0.2740    34.4870
    141.000    -0.2700    34.4880
    141.920    -0.2590    34.4910
    143.000    -0.2570    34.4920
    144.040    -0.2500    34.4940
    145.040    -0.2410    34.4970
    146.000    -0.2390    34.5000
    146.960    -0.2390    34.5000
    148.000    -0.2360    34.5010
    149.000    -0.2370    34.5030
    150.000    -0.2380    34.5040
    151.040    -0.2320    34.5060
    152.040    -0.2280    34.5080
    153.000    -0.2260    34.5080
    154.040    -0.2230    34.5100
    155.000    -0.2180    34.5130
    156.000    -0.2130    34.5150
    157.040    -0.2070    34.5180
    158.080    -0.2040    34.5190
    159.040    -0.1930    34.5210
    160.040    -0.1900    34.5230
    161.040    -0.1900    34.5230
    162.040    -0.1900    34.5230
    163.040    -0.1890    34.5230
    163.960    -0.1880    34.5240
# Drift Data
#      C Npts comp n_var
I0   298  104    1     3
#      Pres       Temp   Salinity
    163.880    -0.1710    34.5270
    163.880    -0.1720    34.5280
    163.880    -0.1720    34.5300
    163.880    -0.1720    34.5310
    163.920    -0.1710    34.5300
    163.920    -0.1710    34.5310
    163.960    -0.1700    34.5310
    163.960    -0.1690    34.5310
    163.960    -0.1690    34.5300
    163.960    -0.1690    34.5300
    163.920    -0.1680    34.5300
    163.880    -0.1680    34.5300
    163.880    -0.1680    34.5300
    163.880    -0.1670    34.5310
    163.840    -0.1670    34.5310
    163.840    -0.1660    34.5310
    163.880    -0.1650    34.5310
    163.920    -0.1660    34.5320
    163.960    -0.1680    34.5340
    163.920    -0.1690    34.5350
    163.960    -0.1690    34.5370
    163.960    -0.1670    34.5350
    163.960    -0.1670    34.5360
    163.920    -0.1680    34.5370
    163.880    -0.1680    34.5370
    163.880    -0.1690    34.5380
    163.880    -0.1680    34.5380
    163.880    -0.1680    34.5380
    163.920    -0.1680    34.5380
    163.960    -0.1680    34.5390
    163.960    -0.1680    34.5380
    163.960    -0.1690    34.5380
    163.960    -0.1680    34.5380
    163.960    -0.1680    34.5390
    163.960    -0.1680    34.5380
    163.920    -0.1680    34.5380
    163.920    -0.1690    34.5390
    163.920    -0.1700    34.5390
    163.880    -0.1690    34.5390
    163.880    -0.1690    34.5380
    163.880    -0.1690    34.5390
    163.920    -0.1690    34.5380
    163.920    -0.1680    34.5370
    163.920    -0.1690    34.5370
    163.960    -0.1690    34.5370
    163.960    -0.1690    34.5370
    163.920    -0.1700    34.5380
    163.920    -0.1700    34.5370
    163.880    -0.1680    34.5370
    163.880    -0.1680    34.5380
    163.880    -0.1690    34.5370
    163.880    -0.1680    34.5370
    163.880    -0.1690    34.5370
    163.920    -0.1680    34.5360
    163.960    -0.1690    34.5370
    163.960    -0.1680    34.5360
    163.960    -0.1680    34.5350
    163.960    -0.1670    34.5350
    163.960    -0.1660    34.5350
    163.960    -0.1670    34.5360
    163.920    -0.1670    34.5360
    163.920    -0.1660    34.5350
    163.880    -0.1660    34.5350
    163.920    -0.1680    34.5360
    163.920    -0.1680    34.5350
    163.920    -0.1670    34.5340
    163.960    -0.1680    34.5330
    163.960    -0.1680    34.5340
    163.960    -0.1690    34.5330
    164.000    -0.1690    34.5330
    164.000    -0.1690    34.5330
    163.960    -0.1690    34.5330
    163.960    -0.1690    34.5330
    163.920    -0.1680    34.5330
    163.920    -0.1680    34.5330
    163.920    -0.1680    34.5340
    163.920    -0.1680    34.5320
    163.960    -0.1700    34.5330
    163.960    -0.1700    34.5320
    163.960    -0.1710    34.5320
    163.960    -0.1710    34.5310
    164.000    -0.1710    34.5300
    164.000    -0.1720    34.5290
    163.960    -0.1720    34.5270
    163.960    -0.1720    34.5280
    164.000    -0.1720    34.5270
    163.960    -0.1720    34.5270
    163.960    -0.1720    34.5270
    163.960    -0.1720    34.5270
    164.000    -0.1720    34.5260
    164.000    -0.1740    34.5250
    164.000    -0.1750    34.5250
    164.040    -0.1760    34.5250
    164.040    -0.1790    34.5260
    164.040    -0.1780    34.5250
    164.040    -0.1770    34.5250
    164.040    -0.1760    34.5270
    164.040    -0.1750    34.5290
    164.040    -0.1760    34.5290
    164.040    -0.1760    34.5280
    164.000    -0.1770    34.5260
    164.040    -0.1800    34.5270
    164.040    -0.1850    34.5250
    164.040    -0.1870    34.5240
# Surface Pressure Time Series
#    C npts n_var               Start
S  298    14     6 2020/02/03 00:38:58
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
   124     -1.150     -1.735   -999.000      14 2020/02/03 00:41:02
  1219     -1.080     -1.737   -999.000      14 2020/02/03 00:59:17
  1840     -1.090     -1.736   -999.000      14 2020/02/03 01:09:38
  1845     -1.100     -1.736   -999.000      14 2020/02/03 01:09:43
  1850     -1.080     -1.736   -999.000      14 2020/02/03 01:09:48
  1856     -1.090     -1.735   -999.000      14 2020/02/03 01:09:54
  1861     -1.090     -1.736   -999.000      14 2020/02/03 01:09:59
  1866     -1.090     -1.735   -999.000      14 2020/02/03 01:10:04
  1871     -1.070     -1.736   -999.000      14 2020/02/03 01:10:09
  1876     -1.080     -1.736   -999.000      14 2020/02/03 01:10:14
  1881     -1.080     -1.737   -999.000      14 2020/02/03 01:10:19
  1887     -1.100     -1.736   -999.000      14 2020/02/03 01:10:25
  1892     -1.100     -1.737   -999.000      14 2020/02/03 01:10:30
  1897     -1.090     -1.737   -999.000      14 2020/02/03 01:10:35
# Fall Time Series
#  Cycle npts     Start                Date
F00  298   26 633578017 2020/01/29 01:53:37
# time     Pres phase                Date
     0   -0.040     1 2020/01/29 01:53:37
   430   20.160    10 2020/01/29 02:00:47
   490   31.160     2 2020/01/29 02:01:47
   538   40.080    10 2020/01/29 02:02:35
   669   60.440    10 2020/01/29 02:04:46
   861   80.200    10 2020/01/29 02:07:58
  1069  100.120    10 2020/01/29 02:11:26
  1469  120.040    10 2020/01/29 02:18:06
  2007  140.000    10 2020/01/29 02:27:04
  2647  160.120    10 2020/01/29 02:37:44
  6248  163.840    10 2020/01/29 03:37:45
  9849  163.840    10 2020/01/29 04:37:46
 13450  163.840    10 2020/01/29 05:37:47
 15140  163.840     3 2020/01/29 06:05:57
 17050  163.840    10 2020/01/29 06:37:47
 20651  163.840    10 2020/01/29 07:37:48
 24219  163.880     3 2020/01/29 08:37:16
 24251  163.880    10 2020/01/29 08:37:48
 27852  163.880    10 2020/01/29 09:37:49
 31452  163.920    10 2020/01/29 10:37:49
 33299  163.920     3 2020/01/29 11:08:36
 35177  163.960    10 2020/01/29 11:39:54
 38778  163.920    10 2020/01/29 12:39:55
 42379  163.880    10 2020/01/29 13:39:56
 45980  163.880    10 2020/01/29 14:39:57
 49581  163.880    10 2020/01/29 15:39:58
# Rise Time Series
#  Cycle npts     Start                Date
R00  298   10 634002548 2020/02/02 23:49:08
# time     Pres phase                Date
     0  164.040     7 2020/02/02 23:49:08
  1421  144.840    13 2020/02/03 00:12:49
  1605  124.960    13 2020/02/03 00:15:53
  1831  104.840    13 2020/02/03 00:19:39
  2064   84.560    13 2020/02/03 00:23:32
  2297   64.480    13 2020/02/03 00:27:25
  2511   44.720    13 2020/02/03 00:30:59
  2731   24.920    13 2020/02/03 00:34:39
  2946    4.640    13 2020/02/03 00:38:14
  2989   -0.200     8 2020/02/03 00:38:57
# Pump Time Series
#  Cycle  npts
B00  298  19
#  Pres P_time  Volts    mAmps  Vacuum        phase
#                               Before  After
   31.160   246   11.71     254  662     -9     10
  164.040    30   11.86     338  639     -9      7
  163.960    31   11.89     350  641     -9     13
  163.880    31   12.00     338  644     -9     13
  163.880    31   11.96     326  647     -9     13
  164.040    31   11.98     345  649     -9     13
  163.960    31   11.98     324  652     -9     13
  163.960    31   11.98     325  654     -9     13
  164.040    31   12.00     309  656     -9     13
  163.920    31   11.95     320  659     -9     13
  163.880    31   11.90     323  661     -9     13
  160.480    31   11.87     317  664     -9     13
  150.360    31   11.89     296  666     -9     13
  112.600    31   11.92     290  669     -9     13
  103.120    31   11.92     301  672     -9     13
   74.640    31   11.94     278  675     -9     13
   62.520    31   11.94     287  678     -9     13
   10.280    31   11.90     246  684     -9     13
    0.000   120   11.99     230  694     -9     14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   298  2.2   33   300   500   600   274    60  6240      9     3    1290  120  4.8688
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  298 2020/01/29 01:53:34
# Pcase vacuums
   7.700   6.370
# Oil valve positions
   1.731   0.804   0.671
# Seek attempts and pump time
      3    -66
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED17  298 2020/01/29 15:39:59 9999/99/99 99:99:99
# Pcase vacuums
   6.370   6.370
# Average 
#   P      T      S
 -999.000 -999.000 -999.000
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA17  298 2020/02/02 23:49:05 9999/99/99 99:99:99
# Pcase vacuums
   6.370   6.840
# Alto Engineering: Surface, version 19
#       C Surface Start Date
EX19  298 2020/02/03 00:38:58
# number of packets previous packets
      6              60
# CTD Status pres offset
 0x4e410000  -1.730
# CTD scans at start and end of profile
# Pres    Temp   Salinity Pres    Temp   Salinity
  164.08 -0.1870 34.5230    0.72 -1.7340 31.7470
# Number of Profile bins, missing bins, and raw CTD samples
    651    486   2971
# Difference between GPS and real-time clock
      0
# Pcase vacuums
   6.840   7.000
# Air Pump max current time  max pressure
      45       310     7.533
# Air Valve Positions
   1.239   0.676
# Air Valve Calibrations
   1.147   0.083   0.615
# Oil Valve Positions
   1.731   0.803
# Oil Valve Calibrations
   1.682   0.844
# Encountered Error Codes
#    Cycle Errors
EE19   298      1
#               Date   Code Description
 2020/02/03 01:09:26 0x0C16 SurfaceGps
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  298  4      0
# MC    Pres      Time  Pcase Vacuum
   1     0.000    483   6.440
   3   163.840     25   6.490
   3   163.880     25   6.420
   3   163.920     25   6.370
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  298       197    2020/08/26 02:52:42
Parameter Table Param_Table-V15.txt |
     IFV = 5.0.97.0	(IFV) |
     VER = 15	(version) |
     STY =        1	(Sensor) |
     MSN =      297	(cycle number) |
     MPS =        2	(phase) |
     MPC =        0	(count) |
  MNM[0] =        1	(count) |
  MNM[1] =       12	(count) |
  MNM[2] =        1	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      100	(dBar) |
  DTD[1] =      135	(dBar) |
  DTD[2] =      500	(dBar) |
     MDS =     2050	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        0	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =      100	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  7776000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =      900	(V) |
     PNC =      300	(inHg) |
     PXC =      700	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        3	(summer attempts) |
  ICA[1] =        3	(winter attempts) |
  ICB[0] =        0	(summer cycles) |
  ICB[1] =        0	(winter cycles) |
  IMT[0] =       50	(summer message threshold) |
  IMT[1] =       50	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       10	(shallow limit for test dBar) |
     ICX =       30	(deep limit for test dBar) |
     ICT =    -1.40	(deg C) |
     ICS =       10	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      300	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       10	(summer minutes) |
  ICR[1] =       10	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =      180	(summer secs) |
  ICO[1] =      180	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      297	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       10	(dBar) |
  DSD[1] =       30	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =     1800	(sec) |
  DST[1] =     1800	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      300	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =       20	(dBar) |
     DDT =     3600	(sec) |
  DFT[0] =      300	(sec) |
  DFT[1] =      345	(sec) |
  DFT[2] =      244	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =      900	(sec) |
  DCT[1] =     7200	(sec) |
  DCT[2] =    14400	(sec) |
     DCA =        3	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =    65470	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        1	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =     7200	(sec) |
     DUC =       67	(under damping factor) |
     DPT =       10	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        3	(dBar) |
     DVW =     1800	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =     1800	(sec) |
     PNS =      297	(cycle) |
  PTO[0] =        0	(sec) |
  PTO[1] =   388800	(sec) |
  PTO[2] =   374400	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        1	(0=dis 1=en) |
  PFM[2] =        0	(0=dis 1=en) |
  PFT[0] =        1	(sec) |
  PFT[1] =     1800	(sec) |
  PFT[2] =     3600	(sec) |
  PFD[0] =      100	(dBar) |
  PFD[1] =      150	(dBar) |
  PFD[2] =      300	(dBar) |
  PFV[0] =        1	(sec) |
  PFV[1] =        5	(sec) |
  PFV[2] =       10	(sec) |
     PDT =     3600	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    10800	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     ASN =      297	(cycle number) |
     ARR =        9	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APN =       30	(sec) |
     APF =       90	(sec) |
     ASI =     1000	(ms) |
     ASD =        1	(dBar) |
     ADD =       20	(dBar) |
     ADT =     3600	(sec) |
     ARC =     4500	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =      297	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SDT =        0	(interval) |
     SSA =        0	(0=dis 1=en) |
     SSI =     1000	(ms) |
     SDS =        5	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SAP =     1200	(sec) |
     SPT =       60	(sec) |
     SSP =      120	(sec) |

