# Message Decoder for Solo2, Alto and Deep Solo Floats; v210127
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11010   281   80 000000000233020  2980     0   6 2019/11/09, 08:05:51,  74.0167 -177.7874   4      0 14
SBD 11010   281  279 000000000233020  2981     0   5 2019/11/09, 08:06:12,  74.0511 -177.7824   4      0 13,6,14
SBD 11010   281  340 000000000233020  2982     0   4 2019/11/09, 08:06:52,  74.0511 -177.7824   3      0 13
SBD 11010   281  319 000000000233020  2983     0   3 2019/11/09, 08:07:09,  74.0524 -177.9127   3      0 3,8
SBD 11010   281  318 000000000233020  2984     0   2 2019/11/09, 08:07:28,  74.0167 -177.7874   2      0 1,2
SBD 11010   281  334 000000000233020  2985     0   1 2019/11/09, 08:08:25,  74.0511 -177.7824   2      0 10,11,5,14,0,15
SBD 11010   281  280 000000000233020  2986     0   0 2019/11/09, 08:08:41,  74.0167 -177.7874   2      0 0,4,14,14,9
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      20         1866           84        1950
# Sensor_ID bytes     % packets Description
          0    48  2.57       2         GPS
          1   101  5.41       1    Pressure
          2   205 10.99       1 Temperature
          3   173  9.27       1    Salinity
          4   108  5.79       1        Fall
          5    53  2.84       1        Rise
          6   169  9.06       1        Pump
          8   134  7.18       1     Surface
          9    86  4.61       1  HR/Drift_P
         10    83  4.45       1  HR/Drift_T
         11    83  4.45       1  HR/Drift_S
         13   407 21.81       2  Parameters
         14   150  8.04       5 Engineering
         15    66  3.54       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  281 1 2019/11/04 08:01 +74.01638 -177.72299   10  6 34 42 49  1.2     0
G  281 2 2019/11/09 07:51 +74.02192 -177.74508   70  5 41 44 49  0.6     0
# Profile Data
#      C Npts comp n_var
D0   281  154    1     3
#      Pres      Temp    Salinity
      1.000    -0.2210    30.4300
      1.960    -0.2210    30.4300
      3.040    -0.2210    30.4300
      4.080    -0.2210    30.4300
      5.080    -0.2210    30.4300
      6.040    -0.2210    30.4300
      7.000    -0.2210    30.4300
      7.960    -0.2210    30.4300
      8.960    -0.2200    30.4300
      9.960    -0.2210    30.4300
     11.000    -0.2200    30.4300
     12.040    -0.2200    30.4300
     13.000    -0.2200    30.4300
     13.920    -0.2200    30.4300
     14.920    -0.2200    30.4300
     15.920    -0.2190    30.4310
     16.960    -0.2190    30.4310
     17.960    -0.2190    30.4310
     19.040    -0.2190    30.4310
     19.960    -0.2180    30.4310
     20.920    -0.2190    30.4310
     22.000    -0.2180    30.4320
     23.000    -0.2180    30.4320
     24.000    -0.2180    30.4320
     24.960    -0.2180    30.4320
     26.040    -0.2180    30.4320
     27.080    -0.2180    30.4320
     28.000    -0.2180    30.4320
     28.960    -0.2180    30.4330
     30.040    -0.2180    30.4340
     31.040    -0.2190    30.4360
     31.920    -0.2230    30.4510
     32.920    -0.2490    30.7110
     34.040    -0.4690    31.4100
     35.000    -0.6620    31.6000
     35.960    -0.6340    31.6260
     37.000    -0.6060    31.6570
     37.960    -0.6210    31.6770
     39.000    -0.6270    31.6960
     40.000    -0.6290    31.7130
     40.920    -0.6340    31.7320
     41.960    -0.6750    31.7480
     43.000    -0.7360    31.7620
     44.000    -0.7920    31.7870
     44.920    -0.8470    31.8000
     45.960    -0.8990    31.8200
     47.040    -0.9210    31.8350
     47.960    -0.9500    31.8540
     48.960    -0.9800    31.8690
     49.960    -1.0410    31.8980
     50.960    -1.1420    31.9260
     51.960    -1.1500    31.9430
     53.000    -1.1590    31.9700
     54.040    -1.1120    31.9930
     55.040    -1.2110    32.0390
     55.920    -1.3270    32.0530
     56.920    -1.2210    32.0550
     57.920    -1.1930    32.0970
     58.920    -1.2320    32.1100
     59.880    -1.2800    32.1280
     60.960    -1.4010    32.1520
     62.040    -1.3220    32.1650
     63.040    -1.2120    32.2150
     64.000    -1.3690    32.2330
     64.960    -1.3740    32.2520
     65.960    -1.3710    32.2870
     66.920    -1.3600    32.3060
     68.000    -1.3310    32.3360
     69.080    -1.3510    32.3620
     70.000    -1.2880    32.3780
     71.000    -1.2310    32.4190
     72.040    -1.1510    32.4310
     73.000    -1.0080    32.4690
     74.040    -0.8480    32.4830
     75.080    -0.7300    32.5110
     76.040    -1.1180    32.5420
     77.000    -1.2360    32.5310
     78.000    -1.2230    32.5500
     79.040    -1.0970    32.5780
     80.040    -1.0700    32.6000
     81.000    -1.0100    32.6130
     82.000    -0.8960    32.6370
     83.000    -0.8140    32.6520
     84.000    -0.3940    32.6650
     85.000    -0.1440    32.7090
     85.960     0.0370    32.7170
     86.960     0.0610    32.7440
     88.040    -0.0320    32.7390
     89.080    -0.1930    32.7540
     90.000    -0.3850    32.7510
     90.960    -0.5470    32.7630
     92.000    -0.7090    32.7810
     93.040    -0.7170    32.8050
     94.040    -0.7420    32.8430
     94.960    -0.6640    32.8700
     95.960    -0.6710    32.9340
     96.960    -0.8780    32.9670
     98.000    -0.9390    32.9880
     99.000    -0.9280    33.0150
    100.040    -0.9110    33.0570
    101.000    -0.9180    33.1130
    101.920    -0.9250    33.1540
    103.000    -1.0580    33.2510
    104.000    -1.0000    33.4450
    105.000    -0.9120    33.5800
    106.040    -0.8610    33.6890
    107.040    -0.7960    33.7960
    108.000    -0.7250    33.8890
    109.000    -0.6640    33.9560
    110.000    -0.6190    34.0090
    111.040    -0.5630    34.0580
    112.040    -0.5230    34.0900
    112.960    -0.4380    34.1600
    114.000    -0.4080    34.1870
    115.040    -0.3670    34.2080
    115.920    -0.3180    34.2460
    116.920    -0.2880    34.2710
    118.000    -0.2260    34.3150
    119.000    -0.1740    34.3530
    119.960    -0.1430    34.3810
    121.040    -0.1150    34.3990
    122.040    -0.0750    34.4250
    122.920    -0.0600    34.4380
    123.920    -0.0400    34.4460
    124.960     0.0450    34.4940
    125.960     0.0550    34.5050
    127.000     0.0630    34.5110
    128.000     0.0650    34.5130
    129.000     0.0670    34.5150
    130.000     0.0710    34.5170
    131.000     0.0820    34.5240
    132.040     0.0900    34.5300
    133.000     0.0970    34.5330
    134.000     0.1040    34.5380
    134.960     0.1080    34.5400
    136.000     0.1110    34.5420
    137.040     0.1120    34.5440
    138.080     0.1190    34.5470
    139.120     0.1290    34.5530
    140.040     0.1340    34.5550
    141.000     0.1420    34.5590
    142.040     0.1440    34.5610
    143.000     0.1440    34.5610
    144.000     0.1440    34.5610
    145.040     0.1440    34.5610
    146.000     0.1440    34.5610
    147.040     0.1440    34.5610
    148.080     0.1440    34.5610
    149.000     0.1450    34.5610
    150.040     0.1450    34.5620
    151.040     0.1460    34.5630
    152.000     0.1480    34.5640
    153.000     0.1480    34.5650
    154.280     0.1480    34.5640
# Drift Data
#      C Npts comp n_var
I0   281  108    1     3
#      Pres       Temp   Salinity
    154.880     0.1160    34.5360
    154.920     0.1180    34.5380
    154.880     0.1210    34.5400
    154.920     0.1220    34.5410
    154.920     0.1230    34.5420
    154.920     0.1260    34.5430
    154.920     0.1270    34.5440
    154.920     0.1290    34.5450
    154.960     0.1300    34.5450
    154.960     0.1310    34.5450
    154.960     0.1320    34.5460
    154.960     0.1330    34.5480
    154.960     0.1340    34.5480
    155.000     0.1390    34.5520
    155.000     0.1400    34.5550
    154.960     0.1440    34.5580
    154.960     0.1460    34.5620
    154.960     0.1480    34.5640
    154.600     0.1490    34.5660
    154.440     0.1500    34.5670
    154.440     0.1500    34.5680
    154.440     0.1490    34.5650
    154.480     0.1470    34.5630
    154.480     0.1470    34.5640
    154.480     0.1470    34.5640
    154.520     0.1470    34.5630
    154.520     0.1480    34.5640
    154.520     0.1480    34.5640
    154.560     0.1490    34.5640
    154.560     0.1490    34.5660
    154.560     0.1490    34.5670
    154.560     0.1490    34.5660
    154.520     0.1480    34.5640
    154.520     0.1480    34.5630
    154.480     0.1470    34.5630
    154.480     0.1480    34.5640
    154.480     0.1480    34.5650
    154.480     0.1490    34.5640
    154.480     0.1480    34.5640
    154.480     0.1480    34.5650
    154.480     0.1480    34.5630
    154.480     0.1480    34.5630
    154.520     0.1460    34.5620
    154.520     0.1460    34.5620
    154.520     0.1470    34.5640
    154.560     0.1490    34.5650
    154.560     0.1480    34.5660
    154.560     0.1480    34.5650
    154.560     0.1470    34.5640
    154.520     0.1460    34.5610
    154.520     0.1450    34.5600
    154.520     0.1440    34.5600
    154.480     0.1440    34.5590
    154.520     0.1440    34.5590
    154.520     0.1420    34.5590
    154.520     0.1420    34.5580
    154.560     0.1410    34.5590
    154.560     0.1410    34.5570
    154.520     0.1380    34.5530
    154.520     0.1360    34.5520
    154.560     0.1380    34.5550
    154.520     0.1360    34.5540
    154.520     0.1320    34.5500
    154.520     0.1300    34.5470
    154.520     0.1320    34.5500
    154.520     0.1280    34.5460
    154.480     0.1240    34.5420
    154.520     0.1270    34.5450
    154.520     0.1290    34.5460
    154.480     0.1270    34.5440
    154.520     0.1250    34.5430
    154.480     0.1190    34.5390
    154.520     0.1190    34.5390
    154.520     0.1190    34.5390
    154.480     0.1160    34.5370
    154.480     0.1110    34.5330
    154.480     0.1080    34.5320
    154.520     0.1070    34.5320
    154.440     0.1130    34.5350
    154.480     0.1140    34.5350
    154.480     0.1130    34.5360
    154.520     0.1120    34.5350
    154.480     0.1120    34.5340
    154.440     0.1100    34.5330
    154.440     0.1080    34.5320
    154.400     0.1060    34.5310
    154.400     0.1060    34.5310
    154.360     0.1060    34.5310
    154.360     0.1060    34.5320
    154.320     0.1070    34.5320
    154.320     0.1070    34.5310
    154.320     0.1080    34.5310
    154.320     0.1100    34.5320
    154.360     0.1090    34.5330
    154.360     0.1090    34.5340
    154.360     0.1120    34.5340
    154.360     0.1140    34.5350
    154.320     0.1130    34.5350
    154.360     0.1270    34.5440
    154.320     0.1290    34.5450
    154.320     0.1320    34.5470
    154.320     0.1390    34.5510
    154.280     0.1430    34.5550
    154.320     0.1470    34.5610
    154.320     0.1460    34.5600
    154.400     0.1420    34.5560
    154.440     0.1430    34.5580
    154.440     0.1470    34.5640
# Surface Pressure Time Series
#    C npts n_var               Start
S  281    14     6 2019/11/09 07:29:23
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
   124     -0.010     -0.290   -999.000      14 2019/11/09 07:31:27
  1219     -0.020     -0.277   -999.000      14 2019/11/09 07:49:42
  1316     -0.110     -0.599   -999.000      14 2019/11/09 07:51:19
  1321     -0.080     -0.609   -999.000      14 2019/11/09 07:51:24
  1326     -0.110     -0.663   -999.000      14 2019/11/09 07:51:29
  1332     -0.090     -0.746   -999.000      14 2019/11/09 07:51:35
  1337     -0.070     -0.813   -999.000      14 2019/11/09 07:51:40
  1342     -0.030     -0.873   -999.000      14 2019/11/09 07:51:45
  1347     -0.010     -0.388   -999.000      14 2019/11/09 07:51:50
  1352      0.130     -0.278   -999.000      14 2019/11/09 07:51:55
  1358     -0.020     -0.272   -999.000      14 2019/11/09 07:52:01
  1363     -0.060     -0.315   -999.000      14 2019/11/09 07:52:06
  1368     -0.030     -0.296   -999.000      14 2019/11/09 07:52:11
  1373     -0.040     -0.351   -999.000      14 2019/11/09 07:52:16
# Fall Time Series
#  Cycle npts     Start                Date
F00  281   20 626169713 2019/11/04 08:01:53
# time     Pres phase                Date
     0   -0.080     1 2019/11/04 08:01:53
   397   20.600    10 2019/11/04 08:08:30
   452   31.280     2 2019/11/04 08:09:25
   505   40.600    10 2019/11/04 08:10:18
   636   60.120    10 2019/11/04 08:12:29
  4236   52.560    10 2019/11/04 09:12:29
  7836   52.320    10 2019/11/04 10:12:29
  8041   52.720     3 2019/11/04 10:15:54
  9899  154.840    10 2019/11/04 10:46:52
 13500  154.880    10 2019/11/04 11:46:53
 17101  154.880     3 2019/11/04 12:46:54
 17104  154.880    10 2019/11/04 12:46:57
 20705  154.880    10 2019/11/04 13:46:58
 24306  154.880    10 2019/11/04 14:46:59
 24345  154.880     3 2019/11/04 14:47:38
 27907  154.880    10 2019/11/04 15:47:00
 31508  154.880    10 2019/11/04 16:47:01
 35109  154.880    10 2019/11/04 17:47:02
 38710  154.880    10 2019/11/04 18:47:03
 38790  154.880    10 2019/11/04 18:48:23
# Rise Time Series
#  Cycle npts     Start                Date
R00  281    9 626597362 2019/11/09 06:49:22
# time     Pres phase                Date
     0  154.400     7 2019/11/09 06:49:22
   795  134.720    13 2019/11/09 07:02:37
  1003  114.480    13 2019/11/09 07:06:05
  1254   94.840    13 2019/11/09 07:10:16
  1511   74.560    13 2019/11/09 07:14:33
  1744   54.440    13 2019/11/09 07:18:26
  1964   34.840    13 2019/11/09 07:22:06
  2246   14.880    13 2019/11/09 07:26:48
  2400   -0.120     8 2019/11/09 07:29:22
# Pump Time Series
#  Cycle  npts
B00  281  15
#  Pres P_time  Volts    mAmps  Vacuum        phase
#                               Before  After
   31.280   386   11.97     252  671     -9     10
  154.400    30   11.78     353  653     -9      7
  154.280    31   11.87     350  655     -9     13
  154.360    31   11.92     328  657     -9     13
  154.240    31   11.89     323  660     -9     13
  154.360    31   11.89     318  663     -9     13
  151.440    31   11.88     307  665     -9     13
  113.560    31   11.83     300  668     -9     13
  102.240    31   11.87     296  672     -9     13
   75.920    31   11.81     284  675     -9     13
   65.520    31   11.93     277  678     -9     13
   35.360    31   11.82     268  681     -9     13
   28.920    31   11.83     257  682     -9     13
    4.000    31   11.85     254  684     -9     13
    0.000   120   12.04     219  693     -9     14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   281  2.2   30   150   135   600   156    30  6480      9     3    1290  120  4.9134
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  281 2019/11/04 08:01:50
# Pcase vacuums
   7.270   6.500
# Oil valve positions
   1.729   0.804   0.656
# Seek attempts and pump time
      3     -5
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED17  281 2019/11/04 18:48:24 9999/99/99 99:99:99
# Pcase vacuums
   6.500   6.510
# Average 
#   P      T      S
 -999.000 -999.000 -999.000
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA17  281 2019/11/09 06:49:18 9999/99/99 99:99:99
# Pcase vacuums
   6.500   6.840
# Alto Engineering: Surface, version 19
#       C Surface Start Date
EX19  281 2019/11/09 07:29:23
# number of packets previous packets
      6              17
# CTD Status pres offset
 0x4e410000   0.120
# CTD scans at start and end of profile
# Pres    Temp   Salinity Pres    Temp   Salinity
  154.40  0.1470 34.5650    0.52 -0.2220 30.4300
# Number of Profile bins, missing bins, and raw CTD samples
    651    496   2383
# Difference between GPS and real-time clock
     -3
# Pcase vacuums
   6.840   7.310
# Air Pump max current time  max pressure
       0      1089     7.461
# Air Valve Positions
   1.236   0.655
# Air Valve Calibrations
   1.147   0.083   0.615
# Oil Valve Positions
   1.730   0.803
# Oil Valve Calibrations
   1.682   0.844
# Encountered Error Codes
#    Cycle Errors
EE19   281      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  281  2      0
# MC    Pres      Time  Pcase Vacuum
   1    -0.040    446   6.370
   3    52.680      6   6.550
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  281       197    2019/11/09 08:06:52
Parameter Table Param_Table-V15.txt |
     IFV = 5.0.97.0	(IFV) |
     VER = 15	(version) |
     STY =        1	(Sensor) |
     MSN =      280	(cycle number) |
     MPS =        1	(phase) |
     MPC =        0	(count) |
  MNM[0] =        1	(count) |
  MNM[1] =       12	(count) |
  MNM[2] =        1	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      100	(dBar) |
  DTD[1] =      135	(dBar) |
  DTD[2] =      500	(dBar) |
     MDS =     2050	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        0	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =      100	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  7776000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =      900	(V) |
     PNC =      300	(inHg) |
     PXC =      700	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        3	(summer attempts) |
  ICA[1] =        3	(winter attempts) |
  ICB[0] =        0	(summer cycles) |
  ICB[1] =        0	(winter cycles) |
  IMT[0] =       50	(summer message threshold) |
  IMT[1] =       50	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       10	(shallow limit for test dBar) |
     ICX =       30	(deep limit for test dBar) |
     ICT =    -1.40	(deg C) |
     ICS =       10	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      300	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       10	(summer minutes) |
  ICR[1] =       10	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =      180	(summer secs) |
  ICO[1] =      180	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      280	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       10	(dBar) |
  DSD[1] =       30	(dBar) |
  DSD[2] =       30	(dBar) |
  DST[0] =     1800	(sec) |
  DST[1] =     1800	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      300	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =       20	(dBar) |
     DDT =     3600	(sec) |
  DFT[0] =      300	(sec) |
  DFT[1] =      384	(sec) |
  DFT[2] =      370	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =      900	(sec) |
  DCT[1] =     7200	(sec) |
  DCT[2] =    14400	(sec) |
     DCA =        3	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =    65534	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        1	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =     7200	(sec) |
     DUC =       67	(under damping factor) |
     DPT =       10	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        3	(dBar) |
     DVW =     1800	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =     1800	(sec) |
     PNS =      280	(cycle) |
  PTO[0] =        0	(sec) |
  PTO[1] =   388800	(sec) |
  PTO[2] =   374400	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        1	(0=dis 1=en) |
  PFM[2] =        0	(0=dis 1=en) |
  PFT[0] =        1	(sec) |
  PFT[1] =     1800	(sec) |
  PFT[2] =     3600	(sec) |
  PFD[0] =      100	(dBar) |
  PFD[1] =      150	(dBar) |
  PFD[2] =      300	(dBar) |
  PFV[0] =        1	(sec) |
  PFV[1] =        5	(sec) |
  PFV[2] =       10	(sec) |
     PDT =     3600	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    10800	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     ASN =      280	(cycle number) |
     ARR =        9	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APN =       30	(sec) |
     APF =       90	(sec) |
     ASI =     1000	(ms) |
     ASD =        1	(dBar) |
     ADD =       20	(dBar) |
     ADT =     3600	(sec) |
     ARC =     4500	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =      280	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       10	(min) |
     SET =       10	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SDT =        0	(interval) |
     SSA =        0	(0=dis 1=en) |
     SSI =     1000	(ms) |
     SDS =        5	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SAP =     1200	(sec) |
     SPT =       60	(sec) |
     SSP =      120	(sec) |

