# Message Decoder for Solo2, Alto and Deep Solo Floats; v210127
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11010   223  318 000000000233020  2336     0   7 2019/02/09, 13:51:00,  70.0736 -174.1011   3      0 6,14,14
SBD 11010   223  335 000000000233020  2337     0   6 2019/02/09, 13:51:16,  70.0736 -174.1011   3      0 6
SBD 11010   223  335 000000000233020  2338     0   5 2019/02/09, 13:51:31,  70.0736 -174.1011   3      0 6
SBD 11010   223  340 000000000233020  2339     0   4 2019/02/09, 13:51:45,  70.0736 -174.1011   4      0 13
SBD 11010   223  318 000000000233020  2340     0   3 2019/02/09, 13:51:59,  70.0736 -174.1011   4      0 8,15,13
SBD 11010   223  192 000000000233020  2341     0   2 2019/02/09, 13:52:15,  70.0736 -174.1011   4      0 1,2,3
SBD 11010   223  306 000000000233020  2342     0   1 2019/02/09, 13:52:35,  70.0770 -174.2057   5      0 10,11,5,14,0
SBD 11010   223  289 000000000233020  2343     0   0 2019/02/09, 13:52:51,  70.0736 -174.1011   5      0 0,4,14,14,9
# Delayed Transmission - (8/8) Messages transmitted  11.4 days after surfacing on 2019/01/29 05:24:00 (refer to Surface Pressure Start Time)
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      22         2337           96        2433
# Sensor_ID bytes     % packets Description
          0    48  2.05       2         GPS
          1    56  2.40       1    Pressure
          2    62  2.65       1 Temperature
          3    62  2.65       1    Salinity
          4   118  5.05       1        Fall
          5    43  1.84       1        Rise
          6   826 35.34       3        Pump
          8   161  6.89       1     Surface
          9    85  3.64       1  HR/Drift_P
         10    93  3.98       1  HR/Drift_T
         11   121  5.18       1  HR/Drift_S
         13   407 17.42       2  Parameters
         14   189  8.09       5 Engineering
         15    66  2.82       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  223 1 9999/99/99 99:99 +99.99000 +999.99000    0  0  0  0  0  0.0     0
G  223 2 9999/99/99 99:99 +99.99000 +999.99000    0  0  0  0  0  0.0     0
# Profile Data
#      C Npts comp n_var
D0   223   47    1     3
#      Pres      Temp    Salinity
      3.200    -1.7670    32.2340
      4.160    -1.7670    32.2330
      4.840    -1.7670    32.2330
      5.960    -1.7670    32.2330
      7.000    -1.7670    32.2320
      8.040    -1.7670    32.2320
      9.040    -1.7670    32.2320
      9.920    -1.7670    32.2320
     11.000    -1.7670    32.2330
     12.080    -1.7670    32.2330
     13.040    -1.7670    32.2330
     14.000    -1.7670    32.2330
     15.000    -1.7670    32.2330
     15.960    -1.7670    32.2320
     16.960    -1.7670    32.2330
     17.920    -1.7670    32.2330
     18.960    -1.7670    32.2330
     19.920    -1.7670    32.2330
     20.920    -1.7660    32.2330
     22.000    -1.7660    32.2340
     23.000    -1.7650    32.2350
     23.960    -1.7650    32.2360
     24.960    -1.7650    32.2360
     25.960    -1.7650    32.2360
     26.920    -1.7650    32.2360
     28.000    -1.7640    32.2390
     29.040    -1.7630    32.2410
     30.000    -1.7630    32.2430
     31.000    -1.7610    32.2470
     32.040    -1.7600    32.2510
     33.040    -1.7580    32.2560
     34.040    -1.7570    32.2590
     35.000    -1.7560    32.2620
     36.000    -1.7450    32.2790
     37.040    -1.7040    32.3610
     38.040    -1.6640    32.4590
     39.040    -1.6550    32.4820
     40.000    -1.6480    32.5020
     40.960    -1.6480    32.5030
     41.960    -1.6460    32.5060
     43.000    -1.6410    32.5240
     44.000    -1.6390    32.5420
     44.960    -1.6400    32.5570
     45.920    -1.6420    32.5640
     46.960    -1.6430    32.5690
     48.080    -1.6440    32.5700
     49.080    -1.6460    32.5740
# Drift Data
#      C Npts comp n_var
I0   223  122    1     3
#      Pres       Temp   Salinity
     48.880    -1.7590    32.2780
     48.920    -1.7600    32.2780
     48.880    -1.7590    32.2770
     48.840    -1.7590    32.2760
     48.800    -1.7560    32.2780
     48.760    -1.7590    32.2810
     48.760    -1.7600    32.2820
     48.800    -1.7610    32.2820
     48.840    -1.7620    32.2820
     48.880    -1.7620    32.2820
     48.920    -1.7620    32.2820
     48.960    -1.7610    32.2810
     49.000    -1.7600    32.2820
     49.000    -1.7580    32.2810
     48.960    -1.7580    32.2800
     48.960    -1.7580    32.2810
     48.920    -1.7600    32.2810
     48.920    -1.7590    32.2810
     48.880    -1.7590    32.2800
     48.880    -1.7580    32.2800
     48.920    -1.7600    32.2800
     48.920    -1.7580    32.2800
     48.960    -1.7550    32.2840
     49.000    -1.7540    32.2960
     49.040    -1.7480    32.3070
     49.040    -1.7410    32.3140
     49.040    -1.7380    32.3190
     49.000    -1.7400    32.3210
     49.000    -1.7400    32.3210
     48.960    -1.7440    32.3270
     48.920    -1.7470    32.3270
     48.920    -1.7450    32.3240
     48.920    -1.7450    32.3200
     48.920    -1.7490    32.3170
     48.960    -1.7550    32.3090
     48.960    -1.7550    32.3070
     49.040    -1.7550    32.3070
     49.040    -1.7550    32.3070
     49.000    -1.7540    32.3060
     49.000    -1.7520    32.3070
     48.960    -1.7530    32.3060
     48.880    -1.7540    32.3050
     48.840    -1.7530    32.3040
     48.840    -1.7550    32.3030
     48.840    -1.7580    32.2980
     48.840    -1.7560    32.2980
     48.880    -1.7560    32.2950
     48.920    -1.7550    32.2920
     48.920    -1.7540    32.2920
     48.920    -1.7530    32.2900
     48.920    -1.7520    32.2880
     48.920    -1.7510    32.2820
     48.920    -1.7530    32.2730
     48.880    -1.7560    32.2600
     48.800    -1.7620    32.2470
     48.800    -1.7630    32.2410
     48.800    -1.7620    32.2380
     48.800    -1.7620    32.2330
     48.800    -1.7620    32.2290
     48.840    -1.7620    32.2240
     48.880    -1.7620    32.2220
     48.880    -1.7620    32.2190
     48.880    -1.7620    32.2210
     48.880    -1.7620    32.2200
     48.840    -1.7620    32.2180
     48.800    -1.7620    32.2190
     48.760    -1.7610    32.2150
     48.720    -1.7610    32.2120
     48.720    -1.7620    32.2170
     48.720    -1.7630    32.2270
     48.720    -1.7620    32.2270
     48.760    -1.7620    32.2250
     48.760    -1.7610    32.2220
     48.800    -1.7600    32.2210
     48.800    -1.7580    32.2200
     48.760    -1.7580    32.2170
     48.760    -1.7570    32.2150
     48.720    -1.7550    32.2110
     48.720    -1.7560    32.2040
     48.680    -1.7550    32.2000
     48.720    -1.7560    32.1960
     48.680    -1.7570    32.1940
     48.680    -1.7570    32.1930
     48.680    -1.7580    32.1940
     48.680    -1.7570    32.2020
     48.720    -1.7540    32.2060
     48.720    -1.7520    32.2080
     48.720    -1.7510    32.2100
     48.720    -1.7500    32.2090
     48.680    -1.7490    32.2100
     48.640    -1.7480    32.2100
     48.640    -1.7460    32.2090
     48.600    -1.7450    32.2070
     48.600    -1.7440    32.2050
     48.600    -1.7430    32.2040
     48.640    -1.7380    32.2280
     48.640    -1.7350    32.2360
     48.640    -1.7320    32.2440
     48.680    -1.7330    32.2530
     48.720    -1.7290    32.2990
     48.720    -1.6890    32.3610
     48.680    -1.6760    32.3960
     48.680    -1.6550    32.4820
     48.680    -1.6470    32.5250
     48.680    -1.6710    32.5730
     48.720    -1.6770    32.5910
     48.760    -1.6710    32.5870
     48.760    -1.6670    32.5800
     48.800    -1.6670    32.5780
     48.840    -1.6760    32.5950
     48.880    -1.6830    32.6000
     48.960    -1.6850    32.6030
     48.960    -1.6830    32.6050
     49.000    -1.6800    32.6060
     48.960    -1.6730    32.6070
     49.000    -1.6690    32.6000
     49.000    -1.6690    32.5910
     49.040    -1.6680    32.5930
     49.040    -1.6630    32.5920
     49.080    -1.6560    32.5820
     49.080    -1.6540    32.5800
     49.120    -1.6480    32.5750
# Surface Pressure Time Series
#    C npts n_var               Start
S  223    17     6 2019/01/29 05:24:00
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
   244      3.220     -1.767   -999.000      14 2019/01/29 05:28:04
  1338      3.200     -1.768   -999.000      14 2019/01/29 05:46:18
  2265      3.200     -1.768   -999.000      14 2019/01/29 06:01:45
  2275      3.220     -1.767   -999.000      14 2019/01/29 06:01:55
  2286      3.220     -1.767   -999.000      14 2019/01/29 06:02:06
  2296      3.220     -1.767   -999.000      14 2019/01/29 06:02:16
  2306      3.210     -1.768   -999.000      14 2019/01/29 06:02:26
  2316      3.220     -1.768   -999.000      14 2019/01/29 06:02:36
  2327      3.210     -1.767   -999.000      14 2019/01/29 06:02:47
  2337      3.230     -1.768   -999.000      14 2019/01/29 06:02:57
  2347      3.220     -1.767   -999.000      14 2019/01/29 06:03:07
  2358      3.200     -1.767   -999.000      14 2019/01/29 06:03:18
  2368      3.220     -1.768   -999.000      14 2019/01/29 06:03:28
  2378      3.220     -1.767   -999.000      14 2019/01/29 06:03:38
  2389      3.220     -1.767   -999.000      14 2019/01/29 06:03:49
  2399      3.220     -1.767   -999.000      14 2019/01/29 06:03:59
  2409      3.230     -1.767   -999.000      14 2019/01/29 06:04:09
# Fall Time Series
#  Cycle npts     Start                Date
F00  223   22 601565549 2019/01/23 13:32:29
# time     Pres phase                Date
     0    0.000     1 2019/01/23 13:32:29
   324   10.640    10 2019/01/23 13:37:53
   377   20.440    10 2019/01/23 13:38:46
   408   26.080     2 2019/01/23 13:39:17
   432   30.400    10 2019/01/23 13:39:41
   489   40.200    10 2019/01/23 13:40:38
  4089   48.880    10 2019/01/23 14:40:38
  7691   48.840    10 2019/01/23 15:40:40
 11291   48.800    10 2019/01/23 16:40:40
 11566   48.800     3 2019/01/23 16:45:15
 14893   48.760    10 2019/01/23 17:40:42
 18495   48.760    10 2019/01/23 18:40:44
 20665   48.760     3 2019/01/23 19:16:54
 22564   48.720    10 2019/01/23 19:48:33
 26164   48.720    10 2019/01/23 20:48:33
 29767   48.760     3 2019/01/23 21:48:36
 29770   48.760    10 2019/01/23 21:48:39
 33371   48.760    10 2019/01/23 22:48:40
 36971   48.800    10 2019/01/23 23:48:40
 40571   48.840    10 2019/01/24 00:48:40
 44174   48.880    10 2019/01/24 01:48:43
 46068   48.880    10 2019/01/24 02:20:17
# Rise Time Series
#  Cycle npts     Start                Date
R00  223    7 602047643 2019/01/29 03:27:23
# time     Pres phase                Date
     0   49.120     7 2019/01/29 03:27:23
  1552   39.280    13 2019/01/29 03:53:15
  1650   29.320    13 2019/01/29 03:54:53
  1748   19.280    13 2019/01/29 03:56:31
  1839    9.520    13 2019/01/29 03:58:02
  5443    3.200    13 2019/01/29 04:58:06
  6996    3.280     8 2019/01/29 05:23:59
# Pump Time Series
#  Cycle  npts
B02  223  74
#  Pres P_time  Volts    mAmps  Vacuum        phase
#                               Before  After
    3.240    30   11.75     224  778     -9     13
    3.160    30   11.69     227  781     -9     13
    3.200    31   11.68     223  783     -9     13
    3.160    31   11.76     230  784     -9     13
    3.240    31   11.72     241  787     -9     13
    3.280    31   11.76     235  789     -9     13
    3.200    31   11.73     221  791     -9     13
    3.280    31   11.71     236  793     -9     13
    3.280    31   11.65     238  795     -9     13
    3.280    31   11.69     238  797     -9     13
    3.200    30   11.73     225  800     -9     13
    3.280    31   11.65     225  802     -9     13
    3.200    31   11.65     231  804     -9     13
    3.280    31   11.64     237  806     -9     13
    3.200    31   11.70     237  808     -9     13
    0.000   240   11.75     168  822     -9     14
    3.200    30   11.85     244  714     -9     13
    3.200    30   11.78     234  716     -9     13
    3.120    31   11.85     241  718     -9     13
    3.280    31   11.81     238  721     -9     13
    3.160    31   11.86     230  723     -9     13
    3.280    31   11.79     221  726     -9     13
    3.160    31   11.76     225  727     -9     13
    3.280    31   11.80     230  730     -9     13
    3.240    31   11.74     236  732     -9     13
    3.200    31   11.76     253  734     -9     13
    3.240    30   11.77     243  737     -9     13
    3.200    30   11.80     240  739     -9     13
    3.240    31   11.79     239  741     -9     13
    3.120    31   11.77     235  743     -9     13
    3.120    31   11.82     232  745     -9     13
    3.120    31   11.81     237  748     -9     13
    3.240    31   11.82     222  750     -9     13
    3.120    31   11.74     246  752     -9     13
    3.280    31   11.75     239  754     -9     13
    3.280    31   11.72     220  756     -9     13
    3.200    30   11.75     239  758     -9     13
    3.200    30   11.71     233  761     -9     13
    3.120    31   11.69     232  763     -9     13
    3.120    31   11.75     222  765     -9     13
    3.280    31   11.74     231  768     -9     13
    3.240    31   11.71     232  770     -9     13
    3.240    31   11.67     231  772     -9     13
    3.160    31   11.71     248  774     -9     13
    3.200    31   11.66     240  776     -9     13
   26.080   353   11.88     222  673     -9     10
   49.120    30   11.86     301  647     -9      7
   49.000    31   11.89     287  649     -9     13
   49.160    31   11.84     297  652     -9     13
   49.200    31   11.86     278  655     -9     13
   49.080    31   11.93     275  657     -9     13
   49.120    31   11.91     286  660     -9     13
   49.160    30   11.92     263  662     -9     13
   49.080    31   11.87     269  665     -9     13
   49.160    31   11.88     259  667     -9     13
   49.160    31   11.94     244  669     -9     13
   49.160    31   11.92     259  672     -9     13
   49.040    31   11.86     271  674     -9     13
   49.120    31   11.86     267  676     -9     13
   49.160    31   11.84     257  679     -9     13
   49.040    31   11.88     251  681     -9     13
   49.160    30   11.86     271  683     -9     13
    4.520    31   11.86     251  686     -9     13
    3.200    31   11.89     244  688     -9     13
    3.120    31   11.88     247  691     -9     13
    3.160    30   11.86     242  693     -9     13
    3.240    31   11.80     247  696     -9     13
    3.160    31   11.89     245  698     -9     13
    3.280    31   11.90     227  700     -9     13
    3.240    31   11.79     238  702     -9     13
    3.280    31   11.84     243  705     -9     13
    3.240    31   11.79     246  707     -9     13
    3.200    31   11.79     245  710     -9     13
    3.240    31   11.87     229  712     -9     13
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   223  2.2   17   300   500   600   215    60  7200      9     3    1395  120  5.4957
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  223 2019/01/23 13:32:26
# Pcase vacuums
   7.610   6.450
# Oil valve positions
   1.728   0.806   0.665
# Seek attempts and pump time
      3   -135
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED17  223 2019/01/24 02:20:18 2019/01/29 02:26:55
# Pcase vacuums
   6.450   6.450
# Average 
#   P      T      S
 -999.000 -999.000 -999.000
# Seek attempts and pump time
      0      0
# End Pressure
   49.080
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA17  223 2019/01/29 03:27:20 9999/99/99 99:99:99
# Pcase vacuums
   6.450   8.080
# Alto Engineering: Surface, version 19
#       C Surface Start Date
EX19  223 2019/01/29 05:24:00
# number of packets previous packets
      8              90
# CTD Status pres offset
 0x4e410000  -2.430
# CTD scans at start and end of profile
# Pres    Temp   Salinity Pres    Temp   Salinity
   49.08 -1.6470 32.5740    3.16 -1.7670 32.2350
# Number of Profile bins, missing bins, and raw CTD samples
    651    601   7228
# Difference between GPS and real-time clock
      0
# Pcase vacuums
   8.080   7.030
# Air Pump max current time  max pressure
      43       396    13.561
# Air Valve Positions
   1.237   0.664
# Air Valve Calibrations
   1.147   0.083   0.615
# Oil Valve Positions
   1.726   0.806
# Oil Valve Calibrations
   1.682   0.844
# Encountered Error Codes
#    Cycle Errors
EE19   223      3
#               Date   Code Description
 2019/01/29 05:23:48 0x0814 AscendPump
 2019/01/29 05:23:48 0x0A14 AscendDepth
 2019/01/29 06:01:28 0x0C16 SurfaceGps
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  223  5      0
# MC    Pres      Time  Pcase Vacuum
   1     0.000    400   6.450
   3    48.800     46   6.470
   3    48.720     47   6.460
   3    48.760     47   6.460
  12    49.080     11   6.450
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  223       197    2019/02/09 13:51:59
Parameter Table Param_Table-V15.txt |
     IFV = 5.0.97.0	(IFV) |
     VER = 15	(version) |
     STY =        1	(Sensor) |
     MSN =      222	(cycle number) |
     MPS =        2	(phase) |
     MPC =        0	(count) |
  MNM[0] =        1	(count) |
  MNM[1] =       21	(count) |
  MNM[2] =        1	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      100	(dBar) |
  DTD[1] =       40	(dBar) |
  DTD[2] =      500	(dBar) |
     MDS =     1000	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        0	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =       30	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  7776000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =      900	(V) |
     PNC =      300	(inHg) |
     PXC =      700	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        3	(summer attempts) |
  ICA[1] =        3	(winter attempts) |
  ICB[0] =        0	(summer cycles) |
  ICB[1] =        0	(winter cycles) |
  IMT[0] =       50	(summer message threshold) |
  IMT[1] =       50	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       10	(shallow limit for test dBar) |
     ICX =       30	(deep limit for test dBar) |
     ICT =    -1.40	(deg C) |
     ICS =       10	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      300	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       10	(summer minutes) |
  ICR[1] =       10	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =      180	(summer secs) |
  ICO[1] =      180	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      222	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       10	(dBar) |
  DSD[1] =        8	(dBar) |
  DSD[2] =       25	(dBar) |
  DST[0] =     1800	(sec) |
  DST[1] =     1800	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      300	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =       10	(dBar) |
     DDT =     3600	(sec) |
  DFT[0] =      300	(sec) |
  DFT[1] =      380	(sec) |
  DFT[2] =      353	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =      900	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    10800	(sec) |
     DCA =        3	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =    65404	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        1	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =     7200	(sec) |
     DUC =      100	(under damping factor) |
     DPT =       10	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =     1800	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =     1800	(sec) |
     PNS =      222	(cycle) |
  PTO[0] =        0	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   432000	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =  2097152	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        1	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =        1	(sec) |
  PFT[1] =      300	(sec) |
  PFT[2] =     3600	(sec) |
  PFD[0] =      100	(dBar) |
  PFD[1] =       45	(dBar) |
  PFD[2] =      300	(dBar) |
  PFV[0] =        1	(sec) |
  PFV[1] =        5	(sec) |
  PFV[2] =       10	(sec) |
     PDT =     3600	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    10800	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     ASN =      222	(cycle number) |
     ARR =        9	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APN =       30	(sec) |
     APF =       60	(sec) |
     ASI =     1000	(ms) |
     ASD =        1	(dBar) |
     ADD =       10	(dBar) |
     ADT =     3600	(sec) |
     ARC =     4500	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =      222	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       15	(min) |
     SET =       15	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SDT =        0	(interval) |
     SSA =        0	(0=dis 1=en) |
     SSI =     1000	(ms) |
     SDS =       10	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SAP =     1200	(sec) |
     SPT =      150	(sec) |
     SSP =      240	(sec) |

