# Message Decoder for Solo2, Alto and Deep Solo Floats; v210127
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11010   216  276 000000000233020  2312     0   5 2018/12/20, 11:54:30,  69.8555 -170.0584 164      0 6,14
SBD 11010   216  340 000000000233020  2313     0   4 2018/12/20, 11:54:45,  69.8555 -170.0584 160      0 13
SBD 11010   216  318 000000000233020  2314     0   3 2018/12/20, 11:55:00,  69.8496 -169.9558 164      0 8,15,13
SBD 11010   216  278 000000000233020  2315     0   2 2018/12/20, 11:55:14,  69.8496 -169.9558 162      0 1,2,3,14
SBD 11010   216  338 000000000233020  2316     0   1 2018/12/20, 11:55:29,  69.9873 -173.9142   7      0 10,11,5,14
SBD 11010   216  313 000000000233020  2317     0   0 2018/12/20, 11:55:44,  69.9908 -174.0184   6      0 0,4,14,14,9,0
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      20         1791           72        1863
# Sensor_ID bytes     % packets Description
          0    48  2.68       2         GPS
          1    50  2.79       1    Pressure
          2    82  4.58       1 Temperature
          3    66  3.69       1    Salinity
          4   118  6.59       1        Fall
          5    43  2.40       1        Rise
          6   224 12.51       1        Pump
          8   161  8.99       1     Surface
          9    85  4.75       1  HR/Drift_P
         10   153  8.54       1  HR/Drift_T
         11   117  6.53       1  HR/Drift_S
         13   407 22.72       2  Parameters
         14   171  9.55       5 Engineering
         15    66  3.69       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  216 1 2018/12/14 20:23 +70.01200 -173.87637   60  8 28 35 40  1.3     0
G  216 2 2018/12/20 11:38 +70.00055 -173.87970   30  9 33 40 48  0.8     0
# Profile Data
#      C Npts comp n_var
D0   216   51    1     3
#      Pres      Temp    Salinity
      0.960    -1.7490    31.9390
      1.960    -1.7490    31.9390
      3.000    -1.7490    31.9390
      4.000    -1.7490    31.9380
      5.040    -1.7490    31.9390
      6.000    -1.7490    31.9380
      7.000    -1.7490    31.9380
      8.040    -1.7490    31.9390
      9.000    -1.7490    31.9400
     10.000    -1.7490    31.9390
     11.080    -1.7490    31.9400
     11.960    -1.7490    31.9400
     12.960    -1.7490    31.9400
     13.960    -1.7490    31.9410
     14.960    -1.7490    31.9420
     16.000    -1.7490    31.9430
     17.000    -1.7490    31.9440
     18.120    -1.7490    31.9440
     19.040    -1.7490    31.9450
     20.040    -1.7490    31.9450
     21.080    -1.7490    31.9460
     22.040    -1.7490    31.9470
     23.000    -1.7490    31.9480
     24.000    -1.7480    31.9500
     25.000    -1.7470    31.9530
     25.960    -1.7470    31.9540
     27.000    -1.7460    31.9560
     28.000    -1.7450    31.9590
     29.000    -1.7300    31.9630
     29.960    -1.7220    31.9720
     30.960    -1.6760    31.9810
     32.000    -1.4100    31.9960
     33.000    -1.1450    32.0760
     34.000    -1.1000    32.1080
     35.080    -1.1860    32.1150
     36.080    -1.1410    32.1210
     37.040    -1.2270    32.1700
     37.960    -1.2880    32.1540
     39.000    -1.1460    32.1450
     40.000    -0.9320    32.1820
     41.000    -0.8970    32.2010
     41.960    -0.8720    32.2110
     43.000    -0.5900    32.2270
     44.040    -0.3530    32.3110
     45.000    -0.3180    32.3200
     46.000    -0.3070    32.3230
     47.000    -0.3030    32.3240
     48.000    -0.3030    32.3240
     49.000    -0.3020    32.3240
     50.080    -0.3020    32.3240
     51.000    -0.3010    32.3240
# Drift Data
#      C Npts comp n_var
I0   216  122    1     3
#      Pres       Temp   Salinity
     51.120    -1.2880    32.1890
     51.120    -1.2330    32.1860
     51.120    -1.2840    32.2000
     51.120    -1.2390    32.2040
     51.160    -1.2250    32.2080
     51.160    -1.2010    32.2010
     51.200    -1.1920    32.2040
     51.160    -1.1780    32.1930
     51.160    -1.1690    32.1880
     51.160    -1.1750    32.1880
     51.160    -1.2290    32.1970
     51.160    -1.2200    32.2090
     51.120    -1.2210    32.2150
     51.080    -1.2400    32.2180
     51.080    -1.2390    32.2160
     51.080    -1.2290    32.2100
     51.080    -1.2230    32.2130
     51.080    -1.2140    32.2100
     51.120    -1.2510    32.2140
     51.120    -1.3030    32.2160
     51.120    -1.3290    32.2160
     51.080    -1.3120    32.2190
     51.120    -1.2710    32.2200
     51.080    -1.2480    32.2230
     51.080    -1.2120    32.2280
     51.040    -1.1760    32.2360
     51.040    -1.1010    32.2520
     51.040    -1.0610    32.2630
     51.000    -1.0700    32.2610
     51.000    -1.1070    32.2530
     51.040    -1.0890    32.2570
     51.040    -1.0750    32.2650
     51.000    -1.0810    32.2640
     51.040    -1.0830    32.2640
     50.960    -1.0930    32.2630
     50.960    -1.0980    32.2590
     50.960    -1.0890    32.2600
     50.920    -1.0690    32.2650
     50.880    -1.0900    32.2640
     50.880    -1.1040    32.2600
     50.880    -1.1100    32.2550
     50.880    -1.0920    32.2500
     50.880    -1.0220    32.2480
     50.880    -1.0050    32.2520
     50.920    -0.9910    32.2630
     50.920    -0.9710    32.2700
     50.920    -0.9790    32.2760
     50.920    -0.9590    32.2770
     50.920    -0.9340    32.2820
     50.880    -0.9070    32.2830
     50.880    -0.8250    32.2900
     50.880    -0.7250    32.3000
     50.880    -0.7050    32.3030
     50.880    -0.6770    32.3090
     50.880    -0.6920    32.3080
     50.880    -0.7590    32.3030
     50.920    -0.7600    32.3030
     50.920    -0.7840    32.3010
     50.920    -0.8420    32.2970
     50.920    -0.8160    32.3000
     50.920    -0.8310    32.2970
     50.920    -0.8470    32.2970
     50.880    -0.9190    32.2900
     50.880    -0.9360    32.2890
     50.840    -0.9420    32.2880
     50.880    -0.9250    32.2890
     50.840    -0.9060    32.2890
     50.880    -0.8550    32.2930
     50.880    -0.8570    32.2940
     50.920    -0.9030    32.2900
     50.960    -0.9040    32.2890
     50.960    -0.8640    32.2930
     50.920    -0.8780    32.2890
     50.920    -0.6950    32.3070
     50.920    -0.6500    32.3110
     50.880    -0.5950    32.3160
     50.880    -0.6450    32.3100
     50.880    -0.7100    32.3020
     50.840    -0.7800    32.2950
     50.880    -0.6480    32.3060
     50.880    -0.6200    32.3090
     50.920    -0.6090    32.3100
     50.920    -0.6120    32.3100
     50.920    -0.6240    32.3100
     50.920    -0.6390    32.3080
     50.920    -0.6500    32.3080
     50.880    -0.6470    32.3070
     50.840    -0.6320    32.3070
     50.840    -0.6200    32.3080
     50.840    -0.6000    32.3090
     50.840    -0.5880    32.3090
     50.840    -0.5810    32.3080
     50.880    -0.5830    32.3060
     50.880    -0.5860    32.3040
     50.920    -0.5880    32.3020
     50.920    -0.5910    32.3000
     50.960    -0.5950    32.2960
     50.920    -0.6030    32.2910
     50.920    -0.5990    32.2880
     50.880    -0.5650    32.2900
     50.880    -0.5540    32.2910
     50.840    -0.5600    32.2900
     50.840    -0.5350    32.2940
     50.840    -0.5130    32.2970
     50.880    -0.5000    32.2980
     50.880    -0.4990    32.2990
     50.920    -0.4960    32.3000
     50.920    -0.4800    32.3020
     50.960    -0.4580    32.3060
     50.960    -0.3710    32.3170
     50.960    -0.3390    32.3200
     50.920    -0.2920    32.3240
     50.920    -0.2820    32.3240
     50.880    -0.2830    32.3230
     50.880    -0.2870    32.3210
     50.880    -0.2870    32.3210
     50.920    -0.2780    32.3240
     50.920    -0.2840    32.3250
     51.000    -0.2950    32.3240
     51.000    -0.2990    32.3250
     51.000    -0.2990    32.3240
     51.040    -0.3010    32.3240
# Surface Pressure Time Series
#    C npts n_var               Start
S  216    17     6 2018/12/20 11:14:52
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
   245     -0.360   -999.000   -999.000      14 2018/12/20 11:18:57
  1339     -0.460   -999.000   -999.000      14 2018/12/20 11:37:11
  1407     -0.460   -999.000   -999.000      14 2018/12/20 11:38:19
  1418     -0.460   -999.000   -999.000      14 2018/12/20 11:38:30
  1428     -0.470   -999.000   -999.000      14 2018/12/20 11:38:40
  1438     -0.470   -999.000   -999.000      14 2018/12/20 11:38:50
  1449     -0.460   -999.000   -999.000      14 2018/12/20 11:39:01
  1459     -0.470   -999.000   -999.000      14 2018/12/20 11:39:11
  1469     -0.450   -999.000   -999.000      14 2018/12/20 11:39:21
  1479     -0.470   -999.000   -999.000      14 2018/12/20 11:39:31
  1490     -0.460   -999.000   -999.000      14 2018/12/20 11:39:42
  1500     -0.450   -999.000   -999.000      14 2018/12/20 11:39:52
  1510     -0.460   -999.000   -999.000      14 2018/12/20 11:40:02
  1521     -0.460   -999.000   -999.000      14 2018/12/20 11:40:13
  1531     -0.470   -999.000   -999.000      14 2018/12/20 11:40:23
  1541     -0.460   -999.000   -999.000      14 2018/12/20 11:40:33
  1552     -0.470   -999.000   -999.000      14 2018/12/20 11:40:44
# Fall Time Series
#  Cycle npts     Start                Date
F00  216   22 598134205 2018/12/14 20:23:25
# time     Pres phase                Date
     0   -0.320     1 2018/12/14 20:23:25
  1809   -0.360     2 2018/12/14 20:53:34
  2952   10.080    10 2018/12/14 21:12:37
  3103   20.040    10 2018/12/14 21:15:08
  3263   30.080    10 2018/12/14 21:17:48
  3413   40.080    10 2018/12/14 21:20:18
  3567   50.160    10 2018/12/14 21:22:52
  7167   51.080    10 2018/12/14 22:22:52
 10770   51.080    10 2018/12/14 23:22:55
 12965   51.080     3 2018/12/14 23:59:30
 14864   51.080    10 2018/12/15 00:31:09
 18464   51.120    10 2018/12/15 01:31:09
 22066   51.120     3 2018/12/15 02:31:11
 22070   51.120    10 2018/12/15 02:31:15
 25671   51.160    10 2018/12/15 03:31:16
 29271   51.160    10 2018/12/15 04:31:16
 31168   51.160     3 2018/12/15 05:02:53
 33067   51.160    10 2018/12/15 05:34:32
 36667   51.160    10 2018/12/15 06:34:32
 40267   51.160    10 2018/12/15 07:34:32
 43870   51.160    10 2018/12/15 08:34:35
 47469   51.120    10 2018/12/15 09:34:34
# Rise Time Series
#  Cycle npts     Start                Date
R00  216    7 598617704 2018/12/20 10:41:44
# time     Pres phase                Date
     0   51.040     7 2018/12/20 10:41:44
  1663   41.400    13 2018/12/20 11:09:27
  1736   31.760    13 2018/12/20 11:10:40
  1816   21.720    13 2018/12/20 11:12:00
  1895   11.680    13 2018/12/20 11:13:19
  1975    1.800    13 2018/12/20 11:14:39
  1987   -0.400     8 2018/12/20 11:14:51
# Pump Time Series
#  Cycle  npts
B00  216  20
#  Pres P_time  Volts    mAmps  Vacuum        phase
#                               Before  After
   -0.360   353   12.12     197  671     -9     10
   51.040    30   11.94     281  641     -9      7
   51.000    31   11.92     295  643     -9     13
   50.920    31   11.99     276  646     -9     13
   51.080    31   11.96     266  649     -9     13
   50.920    31   12.00     265  651     -9     13
   51.080    31   11.91     270  654     -9     13
   51.080    30   11.89     277  656     -9     13
   51.000    31   11.95     266  659     -9     13
   51.040    31   11.97     264  661     -9     13
   51.040    31   11.96     245  663     -9     13
   50.920    31   11.94     259  666     -9     13
   50.920    31   11.87     258  668     -9     13
   50.920    31   11.88     258  671     -9     13
   50.920    31   11.88     261  673     -9     13
   50.920    31   11.92     253  675     -9     13
   51.040    30   11.92     254  678     -9     13
   51.080    31   11.89     252  680     -9     13
   51.040    31   11.93     253  682     -9     13
   -0.360   240   12.02     197  705     -9     14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   216  2.2   17   300   500   600   215    60  7200      9     3    1395  120  5.4957
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  216 2018/12/14 20:23:22
# Pcase vacuums
   7.260   6.430
# Oil valve positions
   1.727   0.804   0.668
# Seek attempts and pump time
      3   -132
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED17  216 2018/12/15 09:34:35 2018/12/20 09:41:16
# Pcase vacuums
   6.430   6.390
# Average 
#   P      T      S
 -999.000 -999.000 -999.000
# Seek attempts and pump time
      0      0
# End Pressure
   51.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA17  216 2018/12/20 10:41:41 9999/99/99 99:99:99
# Pcase vacuums
   6.390   6.870
# Alto Engineering: Surface, version 19
#       C Surface Start Date
EX19  216 2018/12/20 11:14:52
# number of packets previous packets
      6              80
# CTD Status pres offset
 0x4e410000  -0.300
# CTD scans at start and end of profile
# Pres    Temp   Salinity Pres    Temp   Salinity
   51.04 -0.3010 32.3240    0.56 -1.7490 31.9390
# Number of Profile bins, missing bins, and raw CTD samples
    651    599   1968
# Difference between GPS and real-time clock
     -2
# Pcase vacuums
   6.870   7.410
# Air Pump max current time  max pressure
       0      1088     7.426
# Air Valve Positions
   1.236   0.660
# Air Valve Calibrations
   1.147   0.083   0.615
# Oil Valve Positions
   1.725   0.803
# Oil Valve Calibrations
   1.682   0.844
# Encountered Error Codes
#    Cycle Errors
EE19   216      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  216  5      0
# MC    Pres      Time  Pcase Vacuum
   1    -0.360   1803   6.430
   3    51.080     47   6.450
   3    51.120     47   6.440
   3    51.160     46   6.440
  12    51.000     11   6.390
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  216       197    2018/12/20 11:55:00
Parameter Table Param_Table-V15.txt |
     IFV = 5.0.97.0	(IFV) |
     VER = 15	(version) |
     STY =        1	(Sensor) |
     MSN =      215	(cycle number) |
     MPS =        2	(phase) |
     MPC =        0	(count) |
  MNM[0] =        1	(count) |
  MNM[1] =       21	(count) |
  MNM[2] =        1	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      100	(dBar) |
  DTD[1] =       40	(dBar) |
  DTD[2] =      500	(dBar) |
     MDS =     1000	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        0	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =       30	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  7776000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =      900	(V) |
     PNC =      300	(inHg) |
     PXC =      700	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        3	(summer attempts) |
  ICA[1] =        3	(winter attempts) |
  ICB[0] =        0	(summer cycles) |
  ICB[1] =        0	(winter cycles) |
  IMT[0] =       50	(summer message threshold) |
  IMT[1] =       50	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       10	(shallow limit for test dBar) |
     ICX =       30	(deep limit for test dBar) |
     ICT =    -1.40	(deg C) |
     ICS =       10	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      300	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       10	(summer minutes) |
  ICR[1] =       10	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =      180	(summer secs) |
  ICO[1] =      180	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      215	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       10	(dBar) |
  DSD[1] =        8	(dBar) |
  DSD[2] =       25	(dBar) |
  DST[0] =     1800	(sec) |
  DST[1] =     1800	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      300	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =       10	(dBar) |
     DDT =     3600	(sec) |
  DFT[0] =      300	(sec) |
  DFT[1] =      380	(sec) |
  DFT[2] =      353	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =      900	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    10800	(sec) |
     DCA =        3	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =    65404	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        1	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =     7200	(sec) |
     DUC =      100	(under damping factor) |
     DPT =       10	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =     1800	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =     1800	(sec) |
     PNS =      215	(cycle) |
  PTO[0] =        0	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   432000	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =  2097152	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        1	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =        1	(sec) |
  PFT[1] =      300	(sec) |
  PFT[2] =     3600	(sec) |
  PFD[0] =      100	(dBar) |
  PFD[1] =       45	(dBar) |
  PFD[2] =      300	(dBar) |
  PFV[0] =        1	(sec) |
  PFV[1] =        5	(sec) |
  PFV[2] =       10	(sec) |
     PDT =     3600	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    10800	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     ASN =      215	(cycle number) |
     ARR =        9	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APN =       30	(sec) |
     APF =       60	(sec) |
     ASI =     1000	(ms) |
     ASD =        1	(dBar) |
     ADD =       10	(dBar) |
     ADT =     3600	(sec) |
     ARC =     4500	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =      215	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       15	(min) |
     SET =       15	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SDT =        0	(interval) |
     SSA =        0	(0=dis 1=en) |
     SSI =     1000	(ms) |
     SDS =       10	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SAP =     1200	(sec) |
     SPT =      150	(sec) |
     SSP =      240	(sec) |

