# Message Decoder for Solo2, Alto and Deep Solo Floats; v210127
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11010   208  235 000000000233020  2275     0   4 2018/11/23, 22:08:18,  69.8065 -172.5014   3      0 6,14,14
SBD 11010   208  340 000000000233020  2276     0   3 2018/11/23, 22:10:14,  69.8020 -172.3985   3      0 13
SBD 11010   208  318 000000000233020  2277     0   2 2018/11/23, 22:10:48,  69.8020 -172.3985   3      0 8,15,13
SBD 11010   208  286 000000000233020  2278     0   1 2018/11/23, 22:11:06,  69.7687 -172.4122   3      0 5,14,0,1,2,3
SBD 11010   208  322 000000000233020  2279     0   0 2018/11/23, 22:12:11,  69.7687 -172.4122   3      0 0,4,14,14,9,10,11
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      20         1441           60        1501
# Sensor_ID bytes     % packets Description
          0    48  3.33       2         GPS
          1    51  3.54       1    Pressure
          2    74  5.14       1 Temperature
          3    74  5.14       1    Salinity
          4   108  7.49       1        Fall
          5    38  2.64       1        Rise
          6   136  9.44       1        Pump
          8   161 11.17       1     Surface
          9    38  2.64       1  HR/Drift_P
         10    45  3.12       1  HR/Drift_T
         11    45  3.12       1  HR/Drift_S
         13   407 28.24       2  Parameters
         14   150 10.41       5 Engineering
         15    66  4.58       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  208 1 2018/11/22 22:19 +69.72150 -172.00235   90  4 35 39 45  0.0     0
G  208 2 2018/11/23 21:47 +69.79008 -172.35251  230  7 32 39 49  0.9     0
# Profile Data
#      C Npts comp n_var
D0   208   52    1     3
#      Pres      Temp    Salinity
      0.960     1.8550    31.0870
      2.000     1.8550    31.0870
      3.080     1.8550    31.0900
      4.000     1.8550    31.0900
      5.040     1.8550    31.0890
      6.000     1.8540    31.0890
      6.960     1.8550    31.0890
      8.000     1.8550    31.0850
      9.000     1.8550    31.0840
     10.040     1.8570    31.0830
     11.000     1.8570    31.0910
     12.000     1.8560    31.0880
     13.000     1.8550    31.0860
     14.000     1.8550    31.0870
     14.880     1.8550    31.0910
     15.960     1.8540    31.0890
     16.960     1.8600    31.0930
     17.960     1.8570    31.0920
     19.040     1.8530    31.0880
     19.960     1.8540    31.0880
     21.000     1.8560    31.0870
     21.920     1.8580    31.0910
     23.000     1.8530    31.0890
     24.080     1.8520    31.0860
     24.920     1.8520    31.0870
     26.000     1.8560    31.0890
     27.080     1.8550    31.0890
     27.960     1.8600    31.0920
     29.000     1.8600    31.0940
     30.040     1.8590    31.0920
     30.960     1.8580    31.0920
     32.000     1.8600    31.0940
     33.000     1.8610    31.0940
     34.040     1.8600    31.0960
     35.040     1.8540    31.0950
     35.960     1.8600    31.1020
     36.920     1.8720    31.1190
     38.000     2.0370    31.1760
     39.000     3.0170    31.7780
     39.960     3.3460    32.2150
     41.000     3.3050    32.2440
     42.000     3.3100    32.2450
     42.960     3.3120    32.2450
     44.000     3.3110    32.2460
     45.120     3.3170    32.2450
     46.040     3.3190    32.2460
     47.040     3.3200    32.2430
     48.040     3.3200    32.2440
     49.040     3.3220    32.2440
     50.000     3.3250    32.2450
     51.160     3.3270    32.2450
     51.720     3.3290    32.2450
# Drift Data
#      C Npts comp n_var
I0   208   15    1     3
#      Pres       Temp   Salinity
     36.720     2.4910    31.2530
     37.040     2.2480    31.0550
     36.800     2.1270    30.9480
     36.920     2.1660    30.9710
     37.400     2.3120    31.0840
     37.680     2.2040    31.0310
     37.400     2.2080    31.0420
     37.440     2.3270    31.1900
     37.760     2.3140    31.1920
     38.120     2.3250    31.2460
     38.720     2.2950    31.2690
     39.000     2.1590    31.1710
     39.120     2.2140    31.2340
     38.960     2.3830    31.3580
     38.160     2.3750    31.3820
# Surface Pressure Time Series
#    C npts n_var               Start
S  208    17     6 2018/11/23 21:21:22
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
   245      0.200      1.445   -999.000      14 2018/11/23 21:25:27
  1339      0.070      0.224   -999.000      14 2018/11/23 21:43:41
  1608      0.060      0.220   -999.000      14 2018/11/23 21:48:10
  1619      0.140      0.509   -999.000      14 2018/11/23 21:48:21
  1629      0.100      0.794   -999.000      14 2018/11/23 21:48:31
  1639      0.000      0.797   -999.000      14 2018/11/23 21:48:41
  1649      0.020      0.904   -999.000      14 2018/11/23 21:48:51
  1660     -0.010      0.621   -999.000      14 2018/11/23 21:49:02
  1670      0.170      0.250   -999.000      14 2018/11/23 21:49:12
  1680      0.080      0.530   -999.000      14 2018/11/23 21:49:22
  1691      0.000     -0.839   -999.000      14 2018/11/23 21:49:33
  1701      0.170      1.050   -999.000      14 2018/11/23 21:49:43
  1711      0.050      1.398   -999.000      14 2018/11/23 21:49:53
  1722      0.100      0.366   -999.000      14 2018/11/23 21:50:04
  1732     -0.020      1.222   -999.000      14 2018/11/23 21:50:14
  1742      0.000      0.367   -999.000      14 2018/11/23 21:50:24
  1752      0.070      0.809   -999.000      14 2018/11/23 21:50:34
# Fall Time Series
#  Cycle npts     Start                Date
F00  208   20 596240360 2018/11/22 22:19:20
# time     Pres phase                Date
     0    0.040     1 2018/11/22 22:19:20
   366    9.120     2 2018/11/22 22:25:26
   374   10.400    10 2018/11/22 22:25:34
   435   20.280    10 2018/11/22 22:26:35
   497   30.080    10 2018/11/22 22:27:37
   579   40.400    10 2018/11/22 22:28:59
   694   50.360    10 2018/11/22 22:30:54
   998   40.920    10 2018/11/22 22:35:58
  2553   36.160     3 2018/11/22 23:01:53
  4598   35.760    10 2018/11/22 23:35:58
  8201   36.600    10 2018/11/23 00:36:01
  9797   36.400     3 2018/11/23 01:02:37
 11804   36.720    10 2018/11/23 01:36:04
 15406   36.960    10 2018/11/23 02:36:06
 17042   37.240     3 2018/11/23 03:03:22
 19006   37.000    10 2018/11/23 03:36:06
 22606   36.800    10 2018/11/23 04:36:06
 26206   36.960    10 2018/11/23 05:36:06
 29808   37.040    10 2018/11/23 06:36:08
 31488   36.760    10 2018/11/23 07:04:08
# Rise Time Series
#  Cycle npts     Start                Date
R00  208    6 596322142 2018/11/23 21:02:22
# time     Pres phase                Date
     0   46.000     7 2018/11/23 21:02:22
   741   36.760    13 2018/11/23 21:14:43
   907   26.880    13 2018/11/23 21:17:29
  1004   16.400    13 2018/11/23 21:19:06
  1090    6.640    13 2018/11/23 21:20:32
  1139    0.200     8 2018/11/23 21:21:21
# Pump Time Series
#  Cycle  npts
B00  208  12
#  Pres P_time  Volts    mAmps  Vacuum        phase
#                               Before  After
    9.120   383   12.17     194  654     -9     10
   46.000    30   12.14     254  639     -9      7
   51.720    31   12.19     256  640     -9     13
   51.680    31   12.18     251  643     -9     13
   51.920    31   12.23     240  646     -9     13
   51.720    31   12.16     243  648     -9     13
   51.720    30   12.22     238  651     -9     13
   49.520    31   12.19     237  653     -9     13
   44.040    31   12.21     221  656     -9     13
   36.560    31   12.16     225  660     -9     13
   31.880    31   12.22     215  664     -9     13
    0.000   240   12.29     186  684     -9     14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   208  2.2   16    45    40   600    66     5     0      9     3    1395  120  0.3212
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  208 2018/11/22 22:19:17
# Pcase vacuums
   7.350   6.520
# Oil valve positions
   1.725   0.803   0.642
# Seek attempts and pump time
      3      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED17  208 2018/11/23 07:04:09 2018/11/23 21:01:01
# Pcase vacuums
   6.510   6.370
# Average 
#   P      T      S
 -999.000 -999.000 -999.000
# Seek attempts and pump time
      0      0
# End Pressure
   38.200
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA17  208 2018/11/23 21:02:18 9999/99/99 99:99:99
# Pcase vacuums
   6.370   6.650
# Alto Engineering: Surface, version 19
#       C Surface Start Date
EX19  208 2018/11/23 21:21:22
# number of packets previous packets
      5               9
# CTD Status pres offset
 0x4e410000   0.110
# CTD scans at start and end of profile
# Pres    Temp   Salinity Pres    Temp   Salinity
   47.16  3.3120 32.2370    0.52  1.8550 31.0800
# Number of Profile bins, missing bins, and raw CTD samples
    651    598   1121
# Difference between GPS and real-time clock
     -1
# Pcase vacuums
   6.660   7.320
# Air Pump max current time  max pressure
       0      1088     7.416
# Air Valve Positions
   1.230   0.643
# Air Valve Calibrations
   1.147   0.083   0.615
# Oil Valve Positions
   1.725   0.805
# Oil Valve Calibrations
   1.682   0.844
# Encountered Error Codes
#    Cycle Errors
EE19   208      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  208  2      0
# MC    Pres      Time  Pcase Vacuum
   1    -0.040    360   6.270
  12    38.280      6   6.380
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  208       197    2018/11/23 22:10:48
Parameter Table Param_Table-V15.txt |
     IFV = 5.0.97.0	(IFV) |
     VER = 15	(version) |
     STY =        1	(Sensor) |
     MSN =      207	(cycle number) |
     MPS =        1	(phase) |
     MPC =       15	(count) |
  MNM[0] =        1	(count) |
  MNM[1] =       21	(count) |
  MNM[2] =        1	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      100	(dBar) |
  DTD[1] =       40	(dBar) |
  DTD[2] =      500	(dBar) |
     MDS =     1000	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        0	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =       30	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  7776000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =      900	(V) |
     PNC =      300	(inHg) |
     PXC =      700	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        3	(summer attempts) |
  ICA[1] =        3	(winter attempts) |
  ICB[0] =        0	(summer cycles) |
  ICB[1] =        0	(winter cycles) |
  IMT[0] =       50	(summer message threshold) |
  IMT[1] =       50	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       10	(shallow limit for test dBar) |
     ICX =       30	(deep limit for test dBar) |
     ICT =    -1.40	(deg C) |
     ICS =       10	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      300	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       10	(summer minutes) |
  ICR[1] =       10	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =      180	(summer secs) |
  ICO[1] =      180	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      207	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       10	(dBar) |
  DSD[1] =        8	(dBar) |
  DSD[2] =       25	(dBar) |
  DST[0] =     1800	(sec) |
  DST[1] =     1800	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      300	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =       10	(dBar) |
     DDT =     3600	(sec) |
  DFT[0] =      300	(sec) |
  DFT[1] =      380	(sec) |
  DFT[2] =      353	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =      900	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    10800	(sec) |
     DCA =        3	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =        0	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        1	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =     7200	(sec) |
     DUC =      100	(under damping factor) |
     DPT =       10	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =     1800	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =     1800	(sec) |
     PNS =      207	(cycle) |
  PTO[0] =        0	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   432000	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =  2097152	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        1	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =        1	(sec) |
  PFT[1] =      300	(sec) |
  PFT[2] =     3600	(sec) |
  PFD[0] =      100	(dBar) |
  PFD[1] =       45	(dBar) |
  PFD[2] =      300	(dBar) |
  PFV[0] =        1	(sec) |
  PFV[1] =        5	(sec) |
  PFV[2] =       10	(sec) |
     PDT =     3600	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    10800	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     ASN =      207	(cycle number) |
     ARR =        9	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APN =       30	(sec) |
     APF =       60	(sec) |
     ASI =     1000	(ms) |
     ASD =        1	(dBar) |
     ADD =       10	(dBar) |
     ADT =     3600	(sec) |
     ARC =     4500	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =      207	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       15	(min) |
     SET =       15	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SDT =        0	(interval) |
     SSA =        0	(0=dis 1=en) |
     SSI =     1000	(ms) |
     SDS =       10	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SAP =     1200	(sec) |
     SPT =      150	(sec) |
     SSP =      240	(sec) |

