# Message Decoder for Solo2, Alto and Deep Solo Floats; v210127
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11010   199  235 000000000233020  2229     0   4 2018/11/14, 22:06:49,  69.7595 -171.4744   5      0 6,14,14
SBD 11010   199  340 000000000233020  2230     0   3 2018/11/14, 22:07:11,  69.7263 -171.4897   4      0 13
SBD 11010   199  318 000000000233020  2231     0   2 2018/11/14, 22:07:56,  69.7312 -171.5920   3      0 8,15,13
SBD 11010   199  301 000000000233020  2232     0   1 2018/11/14, 22:08:12,  69.7263 -171.4897   3      0 5,14,0,1,2,3
SBD 11010   199  332 000000000233020  2233     0   0 2018/11/14, 22:08:29,  69.7138 -171.4897   3      0 0,4,14,14,9,10,11
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      20         1466           60        1526
# Sensor_ID bytes     % packets Description
          0    48  3.27       2         GPS
          1    50  3.41       1    Pressure
          2    82  5.59       1 Temperature
          3    82  5.59       1    Salinity
          4   108  7.37       1        Fall
          5    38  2.59       1        Rise
          6   136  9.28       1        Pump
          8   161 10.98       1     Surface
          9    39  2.66       1  HR/Drift_P
         10    46  3.14       1  HR/Drift_T
         11    53  3.62       1  HR/Drift_S
         13   407 27.76       2  Parameters
         14   150 10.23       5 Engineering
         15    66  4.50       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  199 1 2018/11/13 22:14 +69.79359 -171.79640   40  4 35 42 47  1.4     0
G  199 2 2018/11/14 21:50 +69.72385 -171.49768   40  5 32 39 47  1.1     0
# Profile Data
#      C Npts comp n_var
D0   199   52    1     3
#      Pres      Temp    Salinity
      1.080     2.8210    30.7070
      2.040     2.8230    30.7080
      3.000     2.8170    30.7070
      3.960     2.8140    30.7070
      5.000     2.8140    30.7060
      6.040     2.8250    30.7090
      7.000     2.8290    30.7090
      8.000     2.8370    30.7090
      9.000     2.8820    30.7230
      9.920     2.9000    30.7270
     11.040     2.9010    30.7270
     11.920     2.9000    30.7260
     13.040     2.9100    30.7290
     13.920     2.9190    30.7360
     15.040     2.9260    30.7400
     16.000     2.9280    30.7430
     16.960     2.9310    30.7440
     18.040     2.9330    30.7460
     19.040     2.9350    30.7490
     20.000     2.9360    30.7510
     21.000     2.9360    30.7550
     22.000     2.9340    30.7580
     22.960     2.9320    30.7610
     24.000     2.9330    30.7620
     25.000     2.9360    30.7630
     26.040     2.9380    30.7640
     26.960     2.9400    30.7650
     28.040     2.9370    30.7670
     28.960     2.9260    30.7680
     30.040     2.9220    30.7680
     31.040     2.9150    30.7710
     32.000     2.9350    30.7770
     33.000     2.9640    30.7920
     34.040     3.0810    30.8180
     35.000     3.7040    31.6350
     36.040     3.5960    32.0910
     36.960     3.4860    32.1280
     38.040     3.3960    32.1680
     39.000     3.3380    32.1910
     40.040     3.2070    32.2330
     40.960     3.1680    32.2410
     42.000     3.1560    32.2440
     43.080     3.1520    32.2450
     44.080     3.1490    32.2460
     45.000     3.1440    32.2460
     45.960     3.1450    32.2460
     47.040     3.1460    32.2460
     48.000     3.1450    32.2470
     49.000     3.1440    32.2470
     49.960     3.1430    32.2470
     50.960     3.1410    32.2470
     51.880     3.1400    32.2480
# Drift Data
#      C Npts comp n_var
I0   199   16    1     3
#      Pres       Temp   Salinity
     35.480     3.1000    30.7870
     35.600     3.0980    30.7760
     35.200     3.0420    30.7710
     35.400     2.9800    30.7700
     35.640     3.8300    31.8300
     35.000     3.0140    30.7800
     34.800     2.9980    30.7930
     34.600     3.0500    30.8280
     34.640     3.4920    31.2390
     34.560     3.4490    31.1170
     34.600     3.4350    31.1110
     34.640     3.5400    31.3340
     34.640     3.3650    31.0590
     34.760     3.3800    31.0900
     34.480     3.2400    30.8870
     35.360     3.6490    31.3510
# Surface Pressure Time Series
#    C npts n_var               Start
S  199    17     6 2018/11/14 21:27:07
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
   245      0.000      2.097   -999.000      14 2018/11/14 21:31:12
  1339     -0.120     -1.566   -999.000      14 2018/11/14 21:49:26
  1413     -0.050     -0.018   -999.000      14 2018/11/14 21:50:40
  1424     -0.120      0.133   -999.000      14 2018/11/14 21:50:51
  1434     -0.050     -0.013   -999.000      14 2018/11/14 21:51:01
  1444     -0.100     -0.322   -999.000      14 2018/11/14 21:51:11
  1454     -0.110     -0.673   -999.000      14 2018/11/14 21:51:21
  1465     -0.100     -0.962   -999.000      14 2018/11/14 21:51:32
  1475     -0.110     -1.209   -999.000      14 2018/11/14 21:51:42
  1485     -0.120     -1.417   -999.000      14 2018/11/14 21:51:52
  1496     -0.050     -1.596   -999.000      14 2018/11/14 21:52:03
  1506     -0.030     -1.730   -999.000      14 2018/11/14 21:52:13
  1516     -0.120     -1.592   -999.000      14 2018/11/14 21:52:23
  1527     -0.120     -1.658   -999.000      14 2018/11/14 21:52:34
  1537     -0.080     -1.756   -999.000      14 2018/11/14 21:52:44
  1547     -0.100     -1.862   -999.000      14 2018/11/14 21:52:54
  1558     -0.090     -1.983   -999.000      14 2018/11/14 21:53:05
# Fall Time Series
#  Cycle npts     Start                Date
F00  199   20 595462501 2018/11/13 22:15:01
# time     Pres phase                Date
     0   -0.040     1 2018/11/13 22:15:01
   325    9.560     2 2018/11/13 22:20:26
   329   10.200    10 2018/11/13 22:20:30
   386   20.280    10 2018/11/13 22:21:27
   448   30.200    10 2018/11/13 22:22:29
   526   40.280    10 2018/11/13 22:23:47
   637   50.120    10 2018/11/13 22:25:38
   963   40.960    10 2018/11/13 22:31:04
  2512   35.920     3 2018/11/13 22:56:53
  4563   35.440    10 2018/11/13 23:31:04
  8163   35.520    10 2018/11/14 00:31:04
  9756   35.720     3 2018/11/14 00:57:37
 11763   35.480    10 2018/11/14 01:31:04
 15365   35.560    10 2018/11/14 02:31:06
 17001   35.600     3 2018/11/14 02:58:22
 18965   35.600    10 2018/11/14 03:31:06
 22565   35.840    10 2018/11/14 04:31:06
 26165   35.720    10 2018/11/14 05:31:06
 29767   35.760    10 2018/11/14 06:31:08
 31448   35.440    10 2018/11/14 06:59:09
# Rise Time Series
#  Cycle npts     Start                Date
R00  199    6 595544903 2018/11/14 21:08:23
# time     Pres phase                Date
     0   46.280     7 2018/11/14 21:08:23
   661   36.520    13 2018/11/14 21:19:24
   821   26.720    13 2018/11/14 21:22:04
   937   16.560    13 2018/11/14 21:24:00
  1047    6.760    13 2018/11/14 21:25:50
  1123    0.080     8 2018/11/14 21:27:06
# Pump Time Series
#  Cycle  npts
B00  199  12
#  Pres P_time  Volts    mAmps  Vacuum        phase
#                               Before  After
    9.560   383   12.10     202  653     -9     10
   46.280    30   12.15     259  641     -9      7
   51.960    31   12.19     249  643     -9     13
   51.800    31   12.22     246  645     -9     13
   51.960    30   12.20     246  648     -9     13
   51.880    31   12.16     239  651     -9     13
   49.800    31   12.19     237  653     -9     13
   44.080    31   12.25     229  656     -9     13
   36.200    31   12.25     210  658     -9     13
   31.080    31   12.18     215  661     -9     13
    8.680    31   12.21     215  664     -9     13
    0.040   240   12.31     185  683     -9     14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   199  2.2   16    45    40   600    66     5     0      9     3    1395  120  0.3212
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  199 2018/11/13 22:14:58
# Pcase vacuums
   7.270   6.500
# Oil valve positions
   1.724   0.804   0.634
# Seek attempts and pump time
      3      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED17  199 2018/11/14 06:59:09 2018/11/14 21:06:05
# Pcase vacuums
   6.490   6.380
# Average 
#   P      T      S
 -999.000 -999.000 -999.000
# Seek attempts and pump time
      0      0
# End Pressure
   34.480
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA17  199 2018/11/14 21:08:20 9999/99/99 99:99:99
# Pcase vacuums
   6.380   6.640
# Alto Engineering: Surface, version 19
#       C Surface Start Date
EX19  199 2018/11/14 21:27:07
# number of packets previous packets
      6               3
# CTD Status pres offset
 0x4e410000  -0.030
# CTD scans at start and end of profile
# Pres    Temp   Salinity Pres    Temp   Salinity
   47.32  3.1490 32.2480    0.64  2.8230 30.7060
# Number of Profile bins, missing bins, and raw CTD samples
    651    598   1100
# Difference between GPS and real-time clock
     -1
# Pcase vacuums
   6.640   7.320
# Air Pump max current time  max pressure
       0      1088     7.411
# Air Valve Positions
   1.231   0.637
# Air Valve Calibrations
   1.147   0.083   0.615
# Oil Valve Positions
   1.724   0.804
# Oil Valve Calibrations
   1.682   0.844
# Encountered Error Codes
#    Cycle Errors
EE19   199      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  199  2      0
# MC    Pres      Time  Pcase Vacuum
   1    -0.040    319   6.240
  12    34.480      6   6.370
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  199       197    2018/11/14 22:07:56
Parameter Table Param_Table-V15.txt |
     IFV = 5.0.97.0	(IFV) |
     VER = 15	(version) |
     STY =        1	(Sensor) |
     MSN =      198	(cycle number) |
     MPS =        1	(phase) |
     MPC =        6	(count) |
  MNM[0] =        1	(count) |
  MNM[1] =       21	(count) |
  MNM[2] =        1	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      100	(dBar) |
  DTD[1] =       40	(dBar) |
  DTD[2] =      500	(dBar) |
     MDS =     1000	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        0	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =       30	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  7776000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =      900	(V) |
     PNC =      300	(inHg) |
     PXC =      700	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        3	(summer attempts) |
  ICA[1] =        3	(winter attempts) |
  ICB[0] =        0	(summer cycles) |
  ICB[1] =        0	(winter cycles) |
  IMT[0] =       50	(summer message threshold) |
  IMT[1] =       50	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       10	(shallow limit for test dBar) |
     ICX =       30	(deep limit for test dBar) |
     ICT =    -1.40	(deg C) |
     ICS =       10	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      300	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       10	(summer minutes) |
  ICR[1] =       10	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =      180	(summer secs) |
  ICO[1] =      180	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      198	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       10	(dBar) |
  DSD[1] =        8	(dBar) |
  DSD[2] =       25	(dBar) |
  DST[0] =     1800	(sec) |
  DST[1] =     1800	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      300	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =       10	(dBar) |
     DDT =     3600	(sec) |
  DFT[0] =      300	(sec) |
  DFT[1] =      380	(sec) |
  DFT[2] =      353	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =      900	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    10800	(sec) |
     DCA =        3	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =        0	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        1	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =     7200	(sec) |
     DUC =      100	(under damping factor) |
     DPT =       10	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =     1800	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =     1800	(sec) |
     PNS =      198	(cycle) |
  PTO[0] =        0	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   432000	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =  2097152	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        1	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =        1	(sec) |
  PFT[1] =      300	(sec) |
  PFT[2] =     3600	(sec) |
  PFD[0] =      100	(dBar) |
  PFD[1] =       45	(dBar) |
  PFD[2] =      300	(dBar) |
  PFV[0] =        1	(sec) |
  PFV[1] =        5	(sec) |
  PFV[2] =       10	(sec) |
     PDT =     3600	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    10800	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     ASN =      198	(cycle number) |
     ARR =        9	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APN =       30	(sec) |
     APF =       60	(sec) |
     ASI =     1000	(ms) |
     ASD =        1	(dBar) |
     ADD =       10	(dBar) |
     ADT =     3600	(sec) |
     ARC =     4500	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =      198	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       15	(min) |
     SET =       15	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SDT =        0	(interval) |
     SSA =        0	(0=dis 1=en) |
     SSI =     1000	(ms) |
     SDS =       10	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SAP =     1200	(sec) |
     SPT =      150	(sec) |
     SSP =      240	(sec) |

