# Message Decoder for Solo2, Alto and Deep Solo Floats; v210127
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11010   195  202 000000000233020  2209     0   4 2018/11/10, 22:00:37,  69.8949 -171.2063  27      0 6,14,14
SBD 11010   195  340 000000000233020  2210     0   3 2018/11/10, 22:00:52,  69.8768 -171.5311   5      0 13
SBD 11010   195  318 000000000233020  2211     0   2 2018/11/10, 22:01:10,  69.8768 -171.5311   6      0 8,15,13
SBD 11010   195  251 000000000233020  2212     0   1 2018/11/10, 22:01:41,  69.8768 -171.5311   7      0 14,0,1,2,3
SBD 11010   195  331 000000000233020  2213     0   0 2018/11/10, 22:03:05,  69.8718 -171.4280   5      0 0,4,14,14,9,10,11,5
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      20         1382           60        1442
# Sensor_ID bytes     % packets Description
          0    48  3.47       2         GPS
          1    48  3.47       1    Pressure
          2    77  5.57       1 Temperature
          3    77  5.57       1    Salinity
          4   108  7.81       1        Fall
          5    38  2.75       1        Rise
          6   103  7.45       1        Pump
          8   161 11.65       1     Surface
          9    37  2.68       1  HR/Drift_P
         10    31  2.24       1  HR/Drift_T
         11    31  2.24       1  HR/Drift_S
         13   407 29.45       2  Parameters
         14   150 10.85       5 Engineering
         15    66  4.78       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  195 1 2018/11/09 22:06 +69.86879 -171.20197   10  5 34 43 48  1.2     0
G  195 2 2018/11/10 21:44 +69.86613 -171.40675  120  4 36 43 49  1.3     0
# Profile Data
#      C Npts comp n_var
D0   195   47    1     3
#      Pres      Temp    Salinity
      1.000     3.4240    30.6750
      2.040     3.4240    30.6750
      3.040     3.4250    30.6750
      4.000     3.4260    30.6750
      5.000     3.4250    30.6750
      6.000     3.4240    30.6750
      7.080     3.4250    30.6750
      8.000     3.4250    30.6740
      8.960     3.4240    30.6750
     10.080     3.4240    30.6750
     10.960     3.4250    30.6750
     12.040     3.4240    30.6750
     13.000     3.4260    30.6740
     13.920     3.4270    30.6740
     15.080     3.4260    30.6750
     16.040     3.4290    30.6750
     17.000     3.4310    30.6770
     18.080     3.4320    30.6780
     19.040     3.4330    30.6780
     19.920     3.4310    30.6830
     20.960     3.4300    30.6850
     21.960     3.4290    30.6890
     23.040     3.4290    30.6920
     24.040     3.4280    30.6910
     25.080     3.4280    30.6930
     26.040     3.4270    30.6940
     26.920     3.4250    30.7010
     27.960     3.4220    30.7100
     29.040     3.4230    30.7040
     30.000     3.4190    30.7190
     30.960     3.4170    30.7240
     31.960     3.4160    30.7330
     32.880     3.4760    30.8110
     34.000     4.1270    31.7380
     35.000     3.9450    32.1410
     35.960     3.5480    32.1720
     37.000     3.3560    32.2280
     38.000     3.2470    32.2550
     39.040     3.1940    32.2630
     40.000     3.1670    32.2670
     41.000     3.1480    32.2700
     42.080     3.1360    32.2710
     43.080     3.1350    32.2710
     44.080     3.1350    32.2710
     45.040     3.1350    32.2710
     46.040     3.1340    32.2710
     47.040     3.1320    32.2720
# Drift Data
#      C Npts comp n_var
I0   195   14    1     3
#      Pres       Temp   Salinity
     44.480     3.0360    32.2870
     44.840     3.0360    32.2870
     44.800     3.0370    32.2870
     45.200     3.0390    32.2870
     45.200     3.0420    32.2880
     45.640     3.0480    32.2850
     45.800     3.0580    32.2820
     46.000     3.0670    32.2790
     45.920     3.0770    32.2770
     46.120     3.0880    32.2750
     46.360     3.1010    32.2730
     46.400     3.1100    32.2730
     46.760     3.1180    32.2720
     47.080     3.1250    32.2720
# Surface Pressure Time Series
#    C npts n_var               Start
S  195    17     6 2018/11/10 21:19:51
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
   244      0.150      3.013   -999.000      14 2018/11/10 21:23:55
  1338      0.040      2.227   -999.000      14 2018/11/10 21:42:09
  1495      0.180      1.264   -999.000      14 2018/11/10 21:44:46
  1506      0.090      1.999   -999.000      14 2018/11/10 21:44:57
  1516      0.040      1.301   -999.000      14 2018/11/10 21:45:07
  1526      0.110      1.271   -999.000      14 2018/11/10 21:45:17
  1536      0.170      2.141   -999.000      14 2018/11/10 21:45:27
  1547      0.160      1.959   -999.000      14 2018/11/10 21:45:38
  1557      0.030      2.676   -999.000      14 2018/11/10 21:45:48
  1567      0.150      2.477   -999.000      14 2018/11/10 21:45:58
  1578      0.070      2.227   -999.000      14 2018/11/10 21:46:09
  1588      0.030      3.040   -999.000      14 2018/11/10 21:46:19
  1598      0.220      3.056   -999.000      14 2018/11/10 21:46:29
  1609      0.130      2.231   -999.000      14 2018/11/10 21:46:40
  1619      0.070      2.398   -999.000      14 2018/11/10 21:46:50
  1629      0.140      2.282   -999.000      14 2018/11/10 21:47:00
  1639      0.100      2.890   -999.000      14 2018/11/10 21:47:10
# Fall Time Series
#  Cycle npts     Start                Date
F00  195   20 595116366 2018/11/09 22:06:06
# time     Pres phase                Date
     0   -0.040     1 2018/11/09 22:06:06
   346    9.280     2 2018/11/09 22:11:52
   353   10.920    10 2018/11/09 22:11:59
   399   20.240    10 2018/11/09 22:12:45
   456   30.640    10 2018/11/09 22:13:42
   534   40.160    10 2018/11/09 22:15:00
  1176   30.960    10 2018/11/09 22:25:42
  2533   30.160     3 2018/11/09 22:48:19
  4776   30.320    10 2018/11/09 23:25:42
  8378   30.400    10 2018/11/10 00:25:44
  9777   29.880     3 2018/11/10 00:49:03
 11632   44.160    10 2018/11/10 01:19:58
 15232   44.640    10 2018/11/10 02:19:58
 18832   44.400     3 2018/11/10 03:19:58
 18835   44.240    10 2018/11/10 03:20:01
 22435   44.600    10 2018/11/10 04:20:01
 26037   44.440    10 2018/11/10 05:20:03
 29637   44.560    10 2018/11/10 06:20:03
 33237   44.440    10 2018/11/10 07:20:03
 33275   44.480    10 2018/11/10 07:20:41
# Rise Time Series
#  Cycle npts     Start                Date
R00  195    6 595199248 2018/11/10 21:07:28
# time     Pres phase                Date
     0   47.320     7 2018/11/10 21:07:28
   243   37.680    13 2018/11/10 21:11:31
   440   27.760    13 2018/11/10 21:14:48
   569   17.320    13 2018/11/10 21:16:57
   667    7.720    13 2018/11/10 21:18:35
   743    0.200     8 2018/11/10 21:19:51
# Pump Time Series
#  Cycle  npts
B00  195   9
#  Pres P_time  Volts    mAmps  Vacuum        phase
#                               Before  After
    9.280   382   12.03     200  654     -9     10
   47.320    30   12.18     263  651     -9      7
   45.760    31   12.21     247  653     -9     13
   40.640    31   12.24     241  656     -9     13
   34.480    31   12.22     236  657     -9     13
   30.600    30   12.19     235  660     -9     13
   23.800    31   12.23     221  662     -9     13
    8.920    31   12.17     209  665     -9     13
    0.040   240   12.20     192  684     -9     14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   195  2.2   16    45    40   600    66     5     0      9     3    1395  120  0.3212
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  195 2018/11/09 22:06:03
# Pcase vacuums
   7.310   6.490
# Oil valve positions
   1.724   0.803   0.638
# Seek attempts and pump time
      3     -1
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED17  195 2018/11/10 07:20:42 9999/99/99 99:99:99
# Pcase vacuums
   6.490   6.490
# Average 
#   P      T      S
 -999.000 -999.000 -999.000
# Seek attempts and pump time
      0      0
# End Pressure
    0.000
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA17  195 2018/11/10 21:07:24 9999/99/99 99:99:99
# Pcase vacuums
   6.490   6.650
# Alto Engineering: Surface, version 19
#       C Surface Start Date
EX19  195 2018/11/10 21:19:51
# number of packets previous packets
      5               0
# CTD Status pres offset
 0x4e410000   0.070
# CTD scans at start and end of profile
# Pres    Temp   Salinity Pres    Temp   Salinity
   47.36  3.1330 32.2710    0.56  3.4250 30.6750
# Number of Profile bins, missing bins, and raw CTD samples
    651    603    720
# Difference between GPS and real-time clock
      2
# Pcase vacuums
   6.660   7.290
# Air Pump max current time  max pressure
       0      1088     7.371
# Air Valve Positions
   1.230   0.637
# Air Valve Calibrations
   1.147   0.083   0.615
# Oil Valve Positions
   1.724   0.803
# Oil Valve Calibrations
   1.682   0.844
# Encountered Error Codes
#    Cycle Errors
EE19   195      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  195  2      0
# MC    Pres      Time  Pcase Vacuum
   1    -0.040    340   6.220
   3    29.920      3   6.440
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  195       197    2018/11/10 22:01:10
Parameter Table Param_Table-V15.txt |
     IFV = 5.0.97.0	(IFV) |
     VER = 15	(version) |
     STY =        1	(Sensor) |
     MSN =      194	(cycle number) |
     MPS =        1	(phase) |
     MPC =        2	(count) |
  MNM[0] =        1	(count) |
  MNM[1] =       21	(count) |
  MNM[2] =        1	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      100	(dBar) |
  DTD[1] =       40	(dBar) |
  DTD[2] =      500	(dBar) |
     MDS =     1000	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        0	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =       30	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  7776000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =      900	(V) |
     PNC =      300	(inHg) |
     PXC =      700	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        3	(summer attempts) |
  ICA[1] =        3	(winter attempts) |
  ICB[0] =        0	(summer cycles) |
  ICB[1] =        0	(winter cycles) |
  IMT[0] =       50	(summer message threshold) |
  IMT[1] =       50	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       10	(shallow limit for test dBar) |
     ICX =       30	(deep limit for test dBar) |
     ICT =    -1.40	(deg C) |
     ICS =       10	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      300	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       10	(summer minutes) |
  ICR[1] =       10	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =      180	(summer secs) |
  ICO[1] =      180	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      194	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       10	(dBar) |
  DSD[1] =        8	(dBar) |
  DSD[2] =       25	(dBar) |
  DST[0] =     1800	(sec) |
  DST[1] =     1800	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      300	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =       10	(dBar) |
     DDT =     3600	(sec) |
  DFT[0] =      300	(sec) |
  DFT[1] =      381	(sec) |
  DFT[2] =      353	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =      900	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =    10800	(sec) |
     DCA =        3	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =    65535	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        1	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =     7200	(sec) |
     DUC =      100	(under damping factor) |
     DPT =       10	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =     1800	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =     1800	(sec) |
     PNS =      194	(cycle) |
  PTO[0] =        0	(sec) |
  PTO[1] =        0	(sec) |
  PTO[2] =   432000	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =  2097152	(24 LSB) |
  PTH[2] =        0	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        1	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =        1	(sec) |
  PFT[1] =      300	(sec) |
  PFT[2] =     3600	(sec) |
  PFD[0] =      100	(dBar) |
  PFD[1] =       45	(dBar) |
  PFD[2] =      300	(dBar) |
  PFV[0] =        1	(sec) |
  PFV[1] =        5	(sec) |
  PFV[2] =       10	(sec) |
     PDT =     3600	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    10800	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     ASN =      194	(cycle number) |
     ARR =        9	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APN =       30	(sec) |
     APF =       60	(sec) |
     ASI =     1000	(ms) |
     ASD =        1	(dBar) |
     ADD =       10	(dBar) |
     ADT =     3600	(sec) |
     ARC =     4500	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =      194	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       15	(min) |
     SET =       15	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SDT =        0	(interval) |
     SSA =        0	(0=dis 1=en) |
     SSI =     1000	(ms) |
     SDS =       10	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SAP =     1200	(sec) |
     SPT =      150	(sec) |
     SSP =      240	(sec) |

