# Message Decoder for Solo2, Alto and Deep Solo Floats; v210127
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11010   192  268 000000000233020  2192    34   5 2018/11/07, 22:10:39,  69.8056 -171.0328   7      0 6,14,14
SBD 11010   192  242 000000000233020  2193     0   6 2018/11/07, 22:11:38,  69.8459 -170.9138   6      0 13
SBD 11010   192  340 000000000233020  2194     0   4 2018/11/07, 22:13:07,  69.8459 -170.9138   4      0 13
SBD 11010   192  308 000000000233020  2195     0   3 2018/11/07, 22:13:24,  69.8459 -170.9138   3      0 8
SBD 11010   192  226 000000000233020  2196     0   2 2018/11/07, 22:13:47,  69.8459 -170.9138   3      0 3,15,13
SBD 11010   192  276 000000000233020  2197     0   1 2018/11/07, 22:14:16,  69.8459 -170.9138   3      0 11,5,14,0,1,2
SBD 11010   192  309 000000000233020  2198     0   0 2018/11/07, 22:14:33,  69.8459 -170.9138   2      0 0,4,14,14,9,10
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      21         1885           84        1969
# Sensor_ID bytes     % packets Description
          0    48  2.55       2         GPS
          1    47  2.49       1    Pressure
          2    75  3.98       1 Temperature
          3    69  3.66       1    Salinity
          4   103  5.46       1        Fall
          5    38  2.02       1        Rise
          6   169  8.97       1        Pump
          8   296 15.70       1     Surface
          9    53  2.81       1  HR/Drift_P
         10    67  3.55       1  HR/Drift_T
         11    67  3.55       1  HR/Drift_S
         13   637 33.79       3  Parameters
         14   150  7.96       5 Engineering
         15    66  3.50       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  192 1 2018/11/06 22:16 +69.77773 -170.60771   70  4 36 39 43  2.3     0
G  192 2 2018/11/07 21:51 +69.83044 -170.92412  110  6 29 39 47  1.0     0
# Profile Data
#      C Npts comp n_var
D0   192   46    1     3
#      Pres      Temp    Salinity
      1.120     3.5510    30.4890
      2.040     3.5530    30.4880
      2.960     3.5520    30.4890
      3.960     3.5540    30.4880
      5.040     3.5540    30.4900
      5.960     3.5560    30.4890
      7.000     3.5550    30.4880
      7.960     3.5560    30.4890
      9.040     3.5540    30.4890
      9.960     3.5590    30.4890
     11.040     3.5590    30.4890
     11.960     3.5580    30.4890
     13.080     3.5600    30.4890
     13.960     3.5590    30.4890
     15.000     3.5590    30.4900
     16.040     3.5620    30.4900
     16.960     3.5630    30.4920
     18.000     3.5630    30.4950
     19.080     3.6730    30.5440
     20.080     3.7840    30.6390
     20.920     3.7830    30.6430
     21.960     3.7810    30.6430
     22.960     3.7790    30.6450
     23.920     3.7800    30.6460
     25.040     3.7900    30.6510
     26.040     3.7980    30.6590
     27.040     3.7980    30.6730
     27.960     3.8960    30.7280
     28.960     4.4170    31.1080
     30.000     5.0590    31.7800
     31.040     5.2350    32.0930
     31.920     4.7390    32.1810
     32.960     4.1870    32.1850
     34.080     3.7320    32.2090
     34.960     3.3130    32.2190
     36.040     3.0390    32.2780
     37.000     3.0240    32.2790
     37.960     3.0270    32.2800
     39.080     3.0270    32.2800
     40.080     3.0270    32.2800
     41.080     3.0280    32.2800
     42.080     3.0310    32.2810
     43.000     3.0310    32.2810
     43.960     3.0310    32.2810
     45.000     3.0290    32.2800
     46.160     3.0260    32.2800
# Drift Data
#      C Npts comp n_var
I0   192   30    1     3
#      Pres       Temp   Salinity
     28.360     4.0050    30.7760
     29.280     3.8980    30.7370
     29.120     4.1930    30.9880
     29.120     4.0280    30.7990
     29.320     4.2310    31.0160
     29.000     3.9780    30.7690
     29.360     4.2750    31.0270
     29.360     4.3230    31.0530
     28.800     4.1090    30.8790
     29.320     4.0940    30.8670
     29.360     4.3740    31.0210
     29.880     4.1010    30.8570
     29.240     4.1060    30.8630
     29.480     4.1270    30.8570
     29.360     4.1480    30.8690
     29.680     4.2720    30.9680
     29.680     4.2080    30.9030
     30.000     4.2440    30.9270
     30.200     4.2980    30.9930
     29.800     4.3520    31.0560
     30.000     4.2350    30.9380
     29.000     4.1400    30.8610
     29.320     4.1750    30.8860
     29.040     4.1390    30.8620
     28.760     4.1760    30.8890
     30.040     4.8800    31.7030
     28.960     4.4620    31.2220
     28.960     4.4820    31.2430
     28.840     4.4390    31.2050
     29.680     4.6840    31.4590
# Surface Pressure Time Series
#    C npts n_var               Start
S  192    32     6 2018/11/07 21:27:32
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
   244      0.160      2.988   -999.000      14 2018/11/07 21:31:36
  1338      0.100      1.486   -999.000      14 2018/11/07 21:49:50
  1482      0.070      1.518   -999.000      14 2018/11/07 21:52:14
  1493      0.210      1.207   -999.000      14 2018/11/07 21:52:25
  1503      0.150      2.138   -999.000      14 2018/11/07 21:52:35
  1513      0.080      1.374   -999.000      14 2018/11/07 21:52:45
  1523      0.150      3.169   -999.000      14 2018/11/07 21:52:55
  1534      0.080      2.372   -999.000      14 2018/11/07 21:53:06
  1544      0.220      1.586   -999.000      14 2018/11/07 21:53:16
  1554      0.160      2.837   -999.000      14 2018/11/07 21:53:26
  1565      0.180      1.895   -999.000      14 2018/11/07 21:53:37
  1575      0.080      1.119   -999.000      14 2018/11/07 21:53:47
  1585      0.110      0.487   -999.000      14 2018/11/07 21:53:57
  1596      0.140     -0.048   -999.000      14 2018/11/07 21:54:08
  1606      0.070      0.037   -999.000      14 2018/11/07 21:54:18
  1616      0.090      0.210   -999.000      14 2018/11/07 21:54:28
  1626      0.220      2.568   -999.000      14 2018/11/07 21:54:38
  1637      0.100      2.006   -999.000      14 2018/11/07 21:54:49
  1647      0.100      1.471   -999.000      14 2018/11/07 21:54:59
  1657      0.130      2.933   -999.000      14 2018/11/07 21:55:09
  1668      0.070      2.694   -999.000      14 2018/11/07 21:55:20
  1678      0.110      3.075   -999.000      14 2018/11/07 21:55:30
  1688      0.080      2.123   -999.000      14 2018/11/07 21:55:40
  1699      0.150      1.549   -999.000      14 2018/11/07 21:55:51
  1709      0.120      1.145   -999.000      14 2018/11/07 21:56:01
  1719      0.140      1.839   -999.000      14 2018/11/07 21:56:11
  1729      0.140      1.187   -999.000      14 2018/11/07 21:56:21
  1740      0.190      0.794   -999.000      14 2018/11/07 21:56:32
  1750      0.070      2.544   -999.000      14 2018/11/07 21:56:42
  1760      0.140      1.897   -999.000      14 2018/11/07 21:56:52
  1771      0.100      2.381   -999.000      14 2018/11/07 21:57:03
  1776      0.190      1.981   -999.000      14 2018/11/07 21:57:08
# Fall Time Series
#  Cycle npts     Start                Date
F00  192   19 594857773 2018/11/06 22:16:13
# time     Pres phase                Date
     0   -0.040     1 2018/11/06 22:16:13
   391    9.720     2 2018/11/06 22:22:44
   394   10.280    10 2018/11/06 22:22:47
   452   20.040    10 2018/11/06 22:23:45
   518   30.320    10 2018/11/06 22:24:51
   604   40.280    10 2018/11/06 22:26:17
  1183   30.960    10 2018/11/06 22:35:56
  2582   30.480     3 2018/11/06 22:59:15
  4783   30.080    10 2018/11/06 23:35:56
  8383   29.800    10 2018/11/07 00:35:56
  9823   29.840     3 2018/11/07 00:59:56
 11983   29.960    10 2018/11/07 01:35:56
 15585   29.360    10 2018/11/07 02:35:58
 17064   29.480     3 2018/11/07 03:00:37
 19185   29.480    10 2018/11/07 03:35:58
 22785   29.560    10 2018/11/07 04:35:58
 26385   29.240    10 2018/11/07 05:35:58
 29988   28.520    10 2018/11/07 06:36:01
 31508   28.400    10 2018/11/07 07:01:21
# Rise Time Series
#  Cycle npts     Start                Date
R00  192    6 594940257 2018/11/07 21:10:57
# time     Pres phase                Date
     0   41.200     7 2018/11/07 21:10:57
   577   31.880    13 2018/11/07 21:20:34
   736   21.840    13 2018/11/07 21:23:13
   859   11.760    13 2018/11/07 21:25:16
   976    1.920    13 2018/11/07 21:27:13
   995    0.240     8 2018/11/07 21:27:32
# Pump Time Series
#  Cycle  npts
B00  192  15
#  Pres P_time  Volts    mAmps  Vacuum        phase
#                               Before  After
    9.720   386   12.30     172  653     -9     10
   41.200    24   12.24     247  638     -9      7
   46.160    25   12.26     246  641     -9     13
   46.280    25   12.21     246  644     -9     13
   46.120    24   12.27     223  646     -9     13
   46.280    25   12.21     230  648     -9     13
   45.640    25   12.29     215  650     -9     13
   43.000    25   12.23     227  652     -9     13
   38.960    25   12.26     221  655     -9     13
   32.840    25   12.24     216  655     -9     13
   27.960    25   12.20     217  656     -9     13
   23.720    25   12.19     201  658     -9     13
   17.880    24   12.23     213  661     -9     13
    7.680    25   12.25     208  663     -9     13
   -0.040   240   12.30     168  682     -9     14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   192  2.2   15    40    30   600    66     5     0      9     3    1545  120  0.3223
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  192 2018/11/06 22:16:10
# Pcase vacuums
   7.370   6.440
# Oil valve positions
   1.723   0.803   0.627
# Seek attempts and pump time
      3      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED17  192 2018/11/07 07:01:21 2018/11/07 21:08:18
# Pcase vacuums
   6.440   6.350
# Average 
#   P      T      S
 -999.000 -999.000 -999.000
# Seek attempts and pump time
      0      0
# End Pressure
   28.880
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA17  192 2018/11/07 21:10:54 9999/99/99 99:99:99
# Pcase vacuums
   6.350   6.630
# Alto Engineering: Surface, version 19
#       C Surface Start Date
EX19  192 2018/11/07 21:27:32
# number of packets previous packets
      6               3
# CTD Status pres offset
 0x4e410000   0.240
# CTD scans at start and end of profile
# Pres    Temp   Salinity Pres    Temp   Salinity
   42.12  3.0450 32.2830    0.76  3.5520 30.4900
# Number of Profile bins, missing bins, and raw CTD samples
    651    604    977
# Difference between GPS and real-time clock
     -1
# Pcase vacuums
   6.630   7.340
# Air Pump max current time  max pressure
       0      1088     7.322
# Air Valve Positions
   1.229   0.632
# Air Valve Calibrations
   1.147   0.083   0.615
# Oil Valve Positions
   1.722   0.801
# Oil Valve Calibrations
   1.682   0.844
# Encountered Error Codes
#    Cycle Errors
EE19   192      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  192  2      0
# MC    Pres      Time  Pcase Vacuum
   1     0.000    384   6.200
  12    28.880      6   6.320
# Upload command:
C mps 1;mnm 1 21;dft 1 383;pth 1 2097152;pto 1 0;pfm 1 1;pfd 1 45;pfv 1 5;dsd 1 8;dtd 1 40;dtd 2 500;dsd 2 25;dft 2 353;dct 2 10800;pfd 2 300;pft 2 3600;pfv 2 10;pth 2 0;pto 2 432000;spt 150;apn 30;apf 60;pdt 3600;MIC 1
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  192       197    2018/11/07 22:13:47
Parameter Table Param_Table-V15.txt |
     IFV = 5.0.97.0	(IFV) |
     VER = 15	(version) |
     STY =        1	(Sensor) |
     MSN =      191	(cycle number) |
     MPS =        2	(phase) |
     MPC =        0	(count) |
  MNM[0] =        1	(count) |
  MNM[1] =        3	(count) |
  MNM[2] =        1	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      100	(dBar) |
  DTD[1] =       40	(dBar) |
  DTD[2] =       30	(dBar) |
     MDS =     1000	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        0	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =       30	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  7776000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =      900	(V) |
     PNC =      300	(inHg) |
     PXC =      700	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        3	(summer attempts) |
  ICA[1] =        3	(winter attempts) |
  ICB[0] =        0	(summer cycles) |
  ICB[1] =        0	(winter cycles) |
  IMT[0] =       50	(summer message threshold) |
  IMT[1] =       50	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       10	(shallow limit for test dBar) |
     ICX =       30	(deep limit for test dBar) |
     ICT =    -1.40	(deg C) |
     ICS =       10	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      300	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       10	(summer minutes) |
  ICR[1] =       10	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =      180	(summer secs) |
  ICO[1] =      180	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      191	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       10	(dBar) |
  DSD[1] =       10	(dBar) |
  DSD[2] =        8	(dBar) |
  DST[0] =     1800	(sec) |
  DST[1] =     1800	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      300	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =       10	(dBar) |
     DDT =     3600	(sec) |
  DFT[0] =      300	(sec) |
  DFT[1] =      310	(sec) |
  DFT[2] =      385	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =      900	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =     1800	(sec) |
     DCA =        3	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =        0	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        1	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =     7200	(sec) |
     DUC =      100	(under damping factor) |
     DPT =       10	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =     1800	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =     1800	(sec) |
     PNS =      191	(cycle) |
  PTO[0] =        0	(sec) |
  PTO[1] =     7200	(sec) |
  PTO[2] =        0	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =  2097152	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =        1	(sec) |
  PFT[1] =      300	(sec) |
  PFT[2] =      300	(sec) |
  PFD[0] =      100	(dBar) |
  PFD[1] =      100	(dBar) |
  PFD[2] =       40	(dBar) |
  PFV[0] =        1	(sec) |
  PFV[1] =        1	(sec) |
  PFV[2] =        5	(sec) |
     PDT =     1800	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    10800	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     ASN =      191	(cycle number) |
     ARR =        9	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APN =       20	(sec) |
     APF =       40	(sec) |
     ASI =     1000	(ms) |
     ASD =        1	(dBar) |
     ADD =       10	(dBar) |
     ADT =     3600	(sec) |
     ARC =     4500	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =      191	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       15	(min) |
     SET =       15	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SDT =        0	(interval) |
     SSA =        0	(0=dis 1=en) |
     SSI =     1000	(ms) |
     SDS =       10	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SAP =     1200	(sec) |
     SPT =      300	(sec) |
     SSP =      240	(sec) |

