# Message Decoder for Solo2, Alto and Deep Solo Floats; v210127
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11010   191  279 000000000233020  2186     0   5 2018/11/06, 22:13:26,  69.7841 -170.6229   2      0 6,14,14
SBD 11010   191  340 000000000233020  2187     0   4 2018/11/06, 22:13:41,  69.7841 -170.6229   3      0 13
SBD 11010   191  308 000000000233020  2188     0   3 2018/11/06, 22:13:57,  69.7841 -170.6229   3      0 8
SBD 11010   191  251 000000000233020  2189     0   2 2018/11/06, 22:14:15,  69.7841 -170.6229   3      0 3,15,13
SBD 11010   191  288 000000000233020  2190     0   1 2018/11/06, 22:14:31,  69.7841 -170.6229   3      0 11,5,14,0,1,2
SBD 11010   191  314 000000000233020  2191     0   0 2018/11/06, 22:14:46,  69.7841 -170.6229   4      0 0,4,14,14,9,10
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      20         1708           72        1780
# Sensor_ID bytes     % packets Description
          0    48  2.81       2         GPS
          1    48  2.81       1    Pressure
          2    86  5.04       1 Temperature
          3    94  5.50       1    Salinity
          4   108  6.32       1        Fall
          5    38  2.22       1        Rise
          6   180 10.54       1        Pump
          8   296 17.33       1     Surface
          9    53  3.10       1  HR/Drift_P
         10    67  3.92       1  HR/Drift_T
         11    67  3.92       1  HR/Drift_S
         13   407 23.83       2  Parameters
         14   150  8.78       5 Engineering
         15    66  3.86       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  191 1 2018/11/05 22:17 +69.69898 -170.24717   10  5 35 40 46  1.0     0
G  191 2 2018/11/06 21:53 +69.77698 -170.59638   20  4 41 44 47  1.7     0
# Profile Data
#      C Npts comp n_var
D0   191   48    1     3
#      Pres      Temp    Salinity
      1.040     3.8170    30.4470
      1.960     3.8180    30.4470
      3.000     3.8200    30.4470
      4.000     3.8190    30.4470
      5.040     3.8220    30.4480
      6.000     3.8230    30.4480
      6.960     3.8240    30.4480
      8.000     3.8230    30.4490
      9.000     3.8230    30.4600
      9.960     3.8290    30.5010
     10.880     3.8450    30.8800
     11.960     3.8390    32.1720
     13.080     3.8320    32.1820
     14.120     3.8170    32.1930
     15.080     3.8010    32.2050
     16.080     3.7930    32.2110
     17.040     3.7860    32.2180
     18.080     3.7780    32.2270
     19.000     3.7750    32.2320
     19.920     3.7780    32.2310
     20.960     3.7850    32.2250
     21.960     3.7970    32.2130
     22.920     3.8120    32.1980
     23.960     3.8310    32.1770
     25.040     3.8510    32.1540
     26.040     3.8760    32.1240
     27.040     3.9110    32.0790
     28.080     3.9840    31.9830
     29.040     4.1880    31.6840
     29.920     4.4920    31.1260
     31.000     4.4350    30.9460
     32.000     3.9930    31.2780
     33.040     3.5470    31.6610
     34.080     3.1780    32.0060
     35.040     3.0210    32.1560
     36.040     2.9800    32.1920
     37.040     2.9540    32.2160
     38.000     2.9300    32.2410
     39.040     2.9180    32.2560
     40.040     2.9150    32.2600
     41.080     2.9430    32.3320
     42.080     2.9730    32.4600
     43.040     2.9570    32.4770
     44.000     2.9390    32.4480
     45.000     2.9280    32.4240
     46.000     2.9230    32.3830
     47.040     2.9210    32.3710
     48.000     2.9190    32.3590
# Drift Data
#      C Npts comp n_var
I0   191   30    1     3
#      Pres       Temp   Salinity
     31.280     4.7290    31.2290
     32.480     4.4120    30.9630
     32.080     4.6380    31.2070
     32.200     4.7180    31.3160
     32.480     4.6890    31.2890
     32.520     4.3230    30.8820
     32.040     4.4890    30.9950
     32.040     4.4030    30.9950
     32.400     4.2260    30.8370
     32.040     4.4730    31.0920
     31.840     4.5330    31.2040
     32.160     4.8650    31.4860
     31.840     4.8840    31.4930
     32.160     4.9070    31.4950
     31.880     4.6940    31.3490
     31.200     4.0440    30.7540
     30.600     4.1550    30.8310
     30.800     4.6900    31.2560
     30.760     4.1670    30.7810
     30.360     4.0270    30.7740
     30.320     4.3780    31.0260
     29.800     3.9940    30.7700
     29.720     4.0860    30.8110
     29.480     4.0930    30.8250
     29.960     3.9890    30.7670
     29.880     3.9230    30.7230
     29.960     4.1130    30.8780
     30.080     4.3520    30.9980
     30.200     4.2840    30.9440
     31.120     5.1070    31.6710
# Surface Pressure Time Series
#    C npts n_var               Start
S  191    32     6 2018/11/06 21:30:31
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
   244      0.000      3.524   -999.000      14 2018/11/06 21:34:35
  1339     -0.050      2.751   -999.000      14 2018/11/06 21:52:50
  1398     -0.110      1.598   -999.000      14 2018/11/06 21:53:49
  1409     -0.090      1.462   -999.000      14 2018/11/06 21:54:00
  1419     -0.040      2.203   -999.000      14 2018/11/06 21:54:10
  1429      0.020      2.274   -999.000      14 2018/11/06 21:54:20
  1439     -0.090      2.157   -999.000      14 2018/11/06 21:54:30
  1450     -0.100      1.961   -999.000      14 2018/11/06 21:54:41
  1460      0.060      3.231   -999.000      14 2018/11/06 21:54:51
  1470     -0.070      3.467   -999.000      14 2018/11/06 21:55:01
  1481     -0.030      3.054   -999.000      14 2018/11/06 21:55:12
  1491      0.030      2.744   -999.000      14 2018/11/06 21:55:22
  1501     -0.030      3.469   -999.000      14 2018/11/06 21:55:32
  1512     -0.040      3.061   -999.000      14 2018/11/06 21:55:43
  1522     -0.100      2.468   -999.000      14 2018/11/06 21:55:53
  1532     -0.080      2.213   -999.000      14 2018/11/06 21:56:03
  1542     -0.070      2.011   -999.000      14 2018/11/06 21:56:13
  1553     -0.060      1.704   -999.000      14 2018/11/06 21:56:24
  1563     -0.040      1.726   -999.000      14 2018/11/06 21:56:34
  1573     -0.100      1.655   -999.000      14 2018/11/06 21:56:44
  1584      0.000      1.509   -999.000      14 2018/11/06 21:56:55
  1594     -0.060      1.419   -999.000      14 2018/11/06 21:57:05
  1604     -0.090      1.348   -999.000      14 2018/11/06 21:57:15
  1615      0.060      1.424   -999.000      14 2018/11/06 21:57:26
  1625     -0.070      1.507   -999.000      14 2018/11/06 21:57:36
  1635     -0.070      1.390   -999.000      14 2018/11/06 21:57:46
  1646     -0.100      1.189   -999.000      14 2018/11/06 21:57:57
  1656     -0.020      0.943   -999.000      14 2018/11/06 21:58:07
  1666     -0.080      0.823   -999.000      14 2018/11/06 21:58:17
  1676      0.070      2.084   -999.000      14 2018/11/06 21:58:27
  1687     -0.090      2.633   -999.000      14 2018/11/06 21:58:38
  1692     -0.100      2.497   -999.000      14 2018/11/06 21:58:43
# Fall Time Series
#  Cycle npts     Start                Date
F00  191   20 594771449 2018/11/05 22:17:29
# time     Pres phase                Date
     0    0.040     1 2018/11/05 22:17:29
   387    9.280     2 2018/11/05 22:23:56
   394   10.560    10 2018/11/05 22:24:03
   448   20.000    10 2018/11/05 22:24:57
   509   30.320    10 2018/11/05 22:25:58
   591   40.360    10 2018/11/05 22:27:20
   698   50.320    10 2018/11/05 22:29:07
   981   40.960    10 2018/11/05 22:33:50
  2578   34.200     3 2018/11/05 23:00:27
  4581   33.960    10 2018/11/05 23:33:50
  8180   33.760    10 2018/11/06 00:33:49
  9819   33.800     3 2018/11/06 01:01:08
 11783   33.720    10 2018/11/06 01:33:52
 15383   33.400    10 2018/11/06 02:33:52
 17060   34.160     3 2018/11/06 03:01:49
 18983   33.720    10 2018/11/06 03:33:52
 22583   32.480    10 2018/11/06 04:33:52
 26185   32.360    10 2018/11/06 05:33:54
 29788   32.040    10 2018/11/06 06:33:57
 31504   31.360    10 2018/11/06 07:02:33
# Rise Time Series
#  Cycle npts     Start                Date
R00  191    6 594853929 2018/11/06 21:12:09
# time     Pres phase                Date
     0   40.720     7 2018/11/06 21:12:09
   638   30.800    13 2018/11/06 21:22:47
   853   20.640    13 2018/11/06 21:26:22
   995   10.840    13 2018/11/06 21:28:44
  1095    0.000    13 2018/11/06 21:30:24
  1101    0.160     8 2018/11/06 21:30:30
# Pump Time Series
#  Cycle  npts
B00  191  16
#  Pres P_time  Volts    mAmps  Vacuum        phase
#                               Before  After
    9.280   386   12.27     189  652     -9     10
   40.720    24   12.19     248  636     -9      7
   44.080    25   12.22     252  639     -9     13
   46.360    25   12.29     234  642     -9     13
   47.960    25   12.28     237  644     -9     13
   48.040    25   12.25     233  647     -9     13
   46.840    25   12.27     229  649     -9     13
   44.600    24   12.26     227  651     -9     13
   40.760    25   12.27     214  653     -9     13
   36.080    25   12.26     222  655     -9     13
   30.400    25   12.24     208  654     -9     13
   28.640    25   12.22     222  656     -9     13
   25.720    25   12.25     211  659     -9     13
   20.320    25   12.27     202  661     -9     13
   15.840    25   12.27     205  663     -9     13
   -0.040   240   12.29     189  681     -9     14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   191  2.2   15    40    30   600    66     5     0      9     3    1545  120  0.3223
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  191 2018/11/05 22:17:26
# Pcase vacuums
   7.350   6.430
# Oil valve positions
   1.722   0.803   0.630
# Seek attempts and pump time
      3      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED17  191 2018/11/06 07:02:33 2018/11/06 21:09:30
# Pcase vacuums
   6.430   6.330
# Average 
#   P      T      S
 -999.000 -999.000 -999.000
# Seek attempts and pump time
      0      0
# End Pressure
   30.280
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA17  191 2018/11/06 21:12:06 9999/99/99 99:99:99
# Pcase vacuums
   6.330   6.630
# Alto Engineering: Surface, version 19
#       C Surface Start Date
EX19  191 2018/11/06 21:30:31
# number of packets previous packets
      6               3
# CTD Status pres offset
 0x4e410000   0.180
# CTD scans at start and end of profile
# Pres    Temp   Salinity Pres    Temp   Salinity
   41.36  3.0560 32.6040    0.76  3.8150 30.4490
# Number of Profile bins, missing bins, and raw CTD samples
    651    602   1076
# Difference between GPS and real-time clock
     -1
# Pcase vacuums
   6.620   7.380
# Air Pump max current time  max pressure
       0      1089     7.264
# Air Valve Positions
   1.228   0.628
# Air Valve Calibrations
   1.147   0.083   0.615
# Oil Valve Positions
   1.723   0.802
# Oil Valve Calibrations
   1.682   0.844
# Encountered Error Codes
#    Cycle Errors
EE19   191      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  191  2      0
# MC    Pres      Time  Pcase Vacuum
   1    -0.080    380   6.200
  12    30.240      6   6.290
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  191       197    2018/11/06 22:14:15
Parameter Table Param_Table-V15.txt |
     IFV = 5.0.97.0	(IFV) |
     VER = 15	(version) |
     STY =        1	(Sensor) |
     MSN =      190	(cycle number) |
     MPS =        2	(phase) |
     MPC =        0	(count) |
  MNM[0] =        1	(count) |
  MNM[1] =        3	(count) |
  MNM[2] =        1	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      100	(dBar) |
  DTD[1] =       40	(dBar) |
  DTD[2] =       30	(dBar) |
     MDS =     1000	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        0	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =       30	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  7776000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =      900	(V) |
     PNC =      300	(inHg) |
     PXC =      700	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        3	(summer attempts) |
  ICA[1] =        3	(winter attempts) |
  ICB[0] =        0	(summer cycles) |
  ICB[1] =        0	(winter cycles) |
  IMT[0] =       50	(summer message threshold) |
  IMT[1] =       50	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       10	(shallow limit for test dBar) |
     ICX =       30	(deep limit for test dBar) |
     ICT =    -1.40	(deg C) |
     ICS =       10	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      300	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       10	(summer minutes) |
  ICR[1] =       10	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =      180	(summer secs) |
  ICO[1] =      180	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      190	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       10	(dBar) |
  DSD[1] =       10	(dBar) |
  DSD[2] =        8	(dBar) |
  DST[0] =     1800	(sec) |
  DST[1] =     1800	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      300	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =       10	(dBar) |
     DDT =     3600	(sec) |
  DFT[0] =      300	(sec) |
  DFT[1] =      310	(sec) |
  DFT[2] =      385	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =      900	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =     1800	(sec) |
     DCA =        3	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =        0	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        1	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =     7200	(sec) |
     DUC =      100	(under damping factor) |
     DPT =       10	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =     1800	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =     1800	(sec) |
     PNS =      190	(cycle) |
  PTO[0] =        0	(sec) |
  PTO[1] =     7200	(sec) |
  PTO[2] =        0	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =  2097152	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =        1	(sec) |
  PFT[1] =      300	(sec) |
  PFT[2] =      300	(sec) |
  PFD[0] =      100	(dBar) |
  PFD[1] =      100	(dBar) |
  PFD[2] =       40	(dBar) |
  PFV[0] =        1	(sec) |
  PFV[1] =        1	(sec) |
  PFV[2] =        5	(sec) |
     PDT =     1800	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    10800	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     ASN =      190	(cycle number) |
     ARR =        9	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APN =       20	(sec) |
     APF =       40	(sec) |
     ASI =     1000	(ms) |
     ASD =        1	(dBar) |
     ADD =       10	(dBar) |
     ADT =     3600	(sec) |
     ARC =     4500	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =      190	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       15	(min) |
     SET =       15	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SDT =        0	(interval) |
     SSA =        0	(0=dis 1=en) |
     SSI =     1000	(ms) |
     SDS =       10	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SAP =     1200	(sec) |
     SPT =      300	(sec) |
     SSP =      240	(sec) |

