# Message Decoder for Solo2, Alto and Deep Solo Floats; v210127
#      SN Cycle Size            IMEI MOMSN MTMSN PID       Date      Time  Latitude Longitude Err Status Sensor IDs
SBD 11010   190  279 000000000233020  2180     0   5 2018/11/05, 22:15:44,  69.7161 -170.2316   4      0 6,14,14
SBD 11010   190  340 000000000233020  2181     0   4 2018/11/05, 22:16:01,  69.7161 -170.2316   4      0 13
SBD 11010   190  308 000000000233020  2182     0   3 2018/11/05, 22:16:17,  69.7161 -170.2316   3      0 8
SBD 11010   190  231 000000000233020  2183     0   2 2018/11/05, 22:16:35,  69.7161 -170.2316   3      0 3,15,13
SBD 11010   190  304 000000000233020  2184     0   1 2018/11/05, 22:16:50,  69.7161 -170.2316   3      0 11,5,14,0,1,2
SBD 11010   190  314 000000000233020  2185     0   0 2018/11/05, 22:17:06,  69.7161 -170.2316   3      0 0,4,14,14,9,10
# Raw packet information
#  Packets Packet_bytes Header_bytes Total_bytes
P0      20         1704           72        1776
# Sensor_ID bytes     % packets Description
          0    48  2.82       2         GPS
          1    50  2.93       1    Pressure
          2    90  5.28       1 Temperature
          3    74  4.34       1    Salinity
          4   108  6.34       1        Fall
          5    48  2.82       1        Rise
          6   180 10.56       1        Pump
          8   296 17.37       1     Surface
          9    53  3.11       1  HR/Drift_P
         10    67  3.93       1  HR/Drift_T
         11    67  3.93       1  HR/Drift_S
         13   407 23.88       2  Parameters
         14   150  8.80       5 Engineering
         15    66  3.87       1     Mission
# GPS Position
#    C Phs Date     Time   Latitude  Longitude Tfix  N Signal   Hdop status
G  190 1 2018/11/04 22:07 +69.64959 -169.95018   10  4 33 41 45  0.0     0
G  190 2 2018/11/05 21:55 +69.69736 -170.24171  130  6 31 38 45  1.3     0
# Profile Data
#      C Npts comp n_var
D0   190   52    1     3
#      Pres      Temp    Salinity
      1.040     3.7480    30.4280
      2.000     3.7470    30.4280
      3.000     3.7470    30.4280
      3.960     3.7470    30.4280
      4.960     3.7480    30.4280
      6.000     3.7480    30.4280
      7.000     3.7490    30.4280
      8.080     3.7490    30.4280
      9.000     3.7470    30.4290
     10.000     3.7460    30.4290
     11.000     3.7360    30.4320
     12.000     3.7340    30.4320
     13.040     3.7330    30.4320
     14.000     3.7320    30.4330
     14.920     3.7320    30.4320
     16.040     3.7320    30.4330
     17.040     3.7320    30.4330
     17.960     3.7290    30.4330
     19.040     3.7280    30.4340
     20.000     3.7270    30.4350
     21.040     3.7290    30.4330
     22.040     3.7260    30.4340
     22.960     3.7240    30.4350
     24.040     3.7240    30.4350
     25.040     3.7250    30.4350
     26.040     3.7230    30.4360
     27.000     3.7230    30.4360
     28.040     3.7230    30.4360
     28.960     3.7230    30.4370
     30.040     3.7230    30.4370
     30.960     3.7250    30.4400
     31.960     3.7370    30.4470
     32.960     3.7570    30.4640
     33.920     3.9070    30.5110
     35.000     5.3320    31.7500
     36.120     5.3240    32.1870
     36.960     4.8980    32.1720
     38.040     4.5460    32.1740
     39.120     3.9350    32.2070
     40.000     3.3570    32.2280
     41.000     3.0420    32.2370
     42.080     3.0920    32.2440
     43.000     3.0480    32.2440
     44.000     2.9820    32.2470
     45.000     2.9420    32.2540
     46.040     2.9130    32.2600
     47.000     2.9020    32.2610
     48.000     2.8920    32.2630
     49.080     2.8900    32.2640
     50.080     2.8880    32.2640
     51.080     2.8870    32.2640
     51.840     2.8860    32.2640
# Drift Data
#      C Npts comp n_var
I0   190   30    1     3
#      Pres       Temp   Salinity
     34.160     4.6550    31.0830
     34.200     4.4600    30.8800
     34.280     4.5660    30.9970
     34.280     4.3330    30.7340
     34.120     4.4080    30.7960
     34.160     4.2000    30.5950
     34.360     4.3980    30.8120
     34.360     4.5030    30.9110
     34.000     4.4850    30.8500
     33.920     4.4460    30.8190
     33.360     4.4080    30.7880
     33.600     5.2800    31.7750
     32.880     4.7530    31.1940
     32.400     4.4510    30.8180
     32.920     4.4490    30.8250
     32.840     4.4740    30.8320
     32.960     4.5890    30.9930
     33.280     4.6990    31.1010
     33.280     4.5740    30.9540
     32.840     4.5340    30.9250
     33.200     4.3870    30.7570
     33.000     4.5040    30.8800
     33.680     4.5960    31.0120
     33.440     4.4190    30.7750
     33.760     4.1440    30.6020
     34.080     4.2500    30.6690
     34.240     4.2940    30.7100
     34.200     4.1230    30.6140
     34.000     4.1590    30.6480
     34.240     4.0320    30.5630
# Surface Pressure Time Series
#    C npts n_var               Start
S  190    32     6 2018/11/05 21:31:00
# time   Pressure       Temp       Psal Mission                Date
#  sec       dBar       degC        PSU   phase                 UTC
   245      0.130      3.621   -999.000      14 2018/11/05 21:35:05
  1339     -0.040      3.066   -999.000      14 2018/11/05 21:53:19
  1508     -0.040      3.058   -999.000      14 2018/11/05 21:56:08
  1519     -0.050      3.347   -999.000      14 2018/11/05 21:56:19
  1529      0.000      3.159   -999.000      14 2018/11/05 21:56:29
  1539      0.040      3.048   -999.000      14 2018/11/05 21:56:39
  1549     -0.050      2.940   -999.000      14 2018/11/05 21:56:49
  1560     -0.040      2.849   -999.000      14 2018/11/05 21:57:00
  1570     -0.050      2.788   -999.000      14 2018/11/05 21:57:10
  1580     -0.010      3.316   -999.000      14 2018/11/05 21:57:20
  1591     -0.050      3.179   -999.000      14 2018/11/05 21:57:31
  1601      0.010      3.078   -999.000      14 2018/11/05 21:57:41
  1611     -0.060      2.965   -999.000      14 2018/11/05 21:57:51
  1622     -0.050      2.959   -999.000      14 2018/11/05 21:58:02
  1632     -0.040      3.212   -999.000      14 2018/11/05 21:58:12
  1642      0.040      3.104   -999.000      14 2018/11/05 21:58:22
  1652      0.050      3.063   -999.000      14 2018/11/05 21:58:32
  1663     -0.050      3.095   -999.000      14 2018/11/05 21:58:43
  1673      0.000      3.006   -999.000      14 2018/11/05 21:58:53
  1683     -0.030      3.554   -999.000      14 2018/11/05 21:59:03
  1694     -0.060      3.523   -999.000      14 2018/11/05 21:59:14
  1704     -0.020      3.308   -999.000      14 2018/11/05 21:59:24
  1714     -0.060      3.205   -999.000      14 2018/11/05 21:59:34
  1725     -0.050      3.077   -999.000      14 2018/11/05 21:59:45
  1735     -0.030      2.952   -999.000      14 2018/11/05 21:59:55
  1745      0.040      2.864   -999.000      14 2018/11/05 22:00:05
  1756      0.020      2.865   -999.000      14 2018/11/05 22:00:16
  1766      0.060      2.830   -999.000      14 2018/11/05 22:00:26
  1776     -0.060      2.796   -999.000      14 2018/11/05 22:00:36
  1786      0.070      2.770   -999.000      14 2018/11/05 22:00:46
  1797      0.090      2.752   -999.000      14 2018/11/05 22:00:57
  1802      0.000      2.771   -999.000      14 2018/11/05 22:01:02
# Fall Time Series
#  Cycle npts     Start                Date
F00  190   20 594684473 2018/11/04 22:07:53
# time     Pres phase                Date
     0    0.000     1 2018/11/04 22:07:53
   371    9.560     2 2018/11/04 22:14:04
   374   10.200    10 2018/11/04 22:14:07
   435   20.560    10 2018/11/04 22:15:08
   497   30.320    10 2018/11/04 22:16:10
   571   40.000    10 2018/11/04 22:17:24
   698   50.080    10 2018/11/04 22:19:31
   977   40.800    10 2018/11/04 22:24:10
  2561   34.560     3 2018/11/04 22:50:34
  4577   34.680    10 2018/11/04 23:24:10
  8177   33.600    10 2018/11/05 00:24:10
  9806   34.080     3 2018/11/05 00:51:19
 11777   33.800    10 2018/11/05 01:24:10
 15380   34.040    10 2018/11/05 02:24:13
 17047   33.200     3 2018/11/05 02:52:00
 18982   33.600    10 2018/11/05 03:24:15
 22585   33.400    10 2018/11/05 04:24:18
 26185   32.920    10 2018/11/05 05:24:18
 29785   34.120    10 2018/11/05 06:24:18
 31490   34.160    10 2018/11/05 06:52:43
# Rise Time Series
#  Cycle npts     Start                Date
R00  190    8 594767467 2018/11/05 21:11:07
# time     Pres phase                Date
     0   41.120     7 2018/11/05 21:11:07
   109   51.280    13 2018/11/05 21:12:56
   644   41.800    13 2018/11/05 21:21:51
   798   31.840    13 2018/11/05 21:24:25
   939   21.480    13 2018/11/05 21:26:46
  1055   11.440    13 2018/11/05 21:28:42
  1171    1.600    13 2018/11/05 21:30:38
  1192    0.240     8 2018/11/05 21:30:59
# Pump Time Series
#  Cycle  npts
B00  190  16
#  Pres P_time  Volts    mAmps  Vacuum        phase
#                               Before  After
    9.560   386   12.16     187  652     -9     10
   41.120    24   12.19     251  632     -9      7
   48.200    25   12.27     239  636     -9     13
   51.800    24   12.27     220  639     -9     13
   51.920    25   12.24     230  642     -9     13
   51.840    25   12.30     234  644     -9     13
   51.920    24   12.29     229  647     -9     13
   51.880    25   12.25     219  649     -9     13
   51.000    25   12.25     232  651     -9     13
   48.000    25   12.20     209  653     -9     13
   42.760    25   12.21     215  655     -9     13
   37.680    25   12.23     212  655     -9     13
   33.040    25   12.27     211  656     -9     13
   29.880    25   12.21     219  658     -9     13
    6.960    25   12.15     215  660     -9     13
    0.000   240   12.33     171  679     -9     14
# Argo Mission Data
#  Cycle Data Miss  Prof  Park  Rise  Park  Prof Drift Ascent Seeks Surface Seek   Cycle
#        Form    #  dbar  dbar   min   min   min   min   cm/s     #     sec  min     day
A2   190  2.2   15    40    30   600    66     5     0      9     3    1545  120  0.3223
# Cal Coeff    P                 T                  S
#          gain  offset      gain  offset       gain  offset
CP           25      10      1000       5      1000       1
# Cal Coeff  Surf  P       Surf  T       Surf S
#          gain  offset      gain  offset      gain  offset
CS          100      10      1000       5      1000       1 
# Engineering Data
# Alto Engineering: Descent
#      C    Descent Start Date
EF17  190 2018/11/04 22:07:50
# Pcase vacuums
   7.350   6.430
# Oil valve positions
   1.722   0.803   0.630
# Seek attempts and pump time
      3      0
# Alto Engineering: Drift
#       C Drift Start Date    End Date
ED17  190 2018/11/05 06:52:44 2018/11/05 21:09:45
# Pcase vacuums
   6.430   6.280
# Average 
#   P      T      S
 -999.000 -999.000 -999.000
# Seek attempts and pump time
      0      0
# End Pressure
   34.280
# Alto Engineering: Ascent
#       C Ascent Start Date   Fall Start Time
EA17  190 2018/11/05 21:11:03 9999/99/99 99:99:99
# Pcase vacuums
   6.280   6.600
# Alto Engineering: Surface, version 19
#       C Surface Start Date
EX19  190 2018/11/05 21:31:00
# number of packets previous packets
      6               2
# CTD Status pres offset
 0x4e410000   0.190
# CTD scans at start and end of profile
# Pres    Temp   Salinity Pres    Temp   Salinity
   42.04  3.3150 32.2450    0.68  3.7470 30.4260
# Number of Profile bins, missing bins, and raw CTD samples
    651    598   1168
# Difference between GPS and real-time clock
     -1
# Pcase vacuums
   6.600   7.360
# Air Pump max current time  max pressure
       0      1088     7.299
# Air Valve Positions
   1.229   0.630
# Air Valve Calibrations
   1.147   0.083   0.615
# Oil Valve Positions
   1.723   0.802
# Oil Valve Calibrations
   1.682   0.844
# Encountered Error Codes
#    Cycle Errors
EE19   190      0
# Alto Engineering: Valve Time Series
#       C  No of points Compression
EV00  190  2      0
# MC    Pres      Time  Pcase Vacuum
   1    -0.080    363   6.190
  12    34.320      6   6.270
# Parameter Dump
#   Cycle  Lines  Transmissin Date
M  190       197    2018/11/05 22:16:35
Parameter Table Param_Table-V15.txt |
     IFV = 5.0.97.0	(IFV) |
     VER = 15	(version) |
     STY =        1	(Sensor) |
     MSN =      189	(cycle number) |
     MPS =        2	(phase) |
     MPC =        0	(count) |
  MNM[0] =        1	(count) |
  MNM[1] =        3	(count) |
  MNM[2] =        1	(count) |
  MNS[0] =        1	(count) |
  MNS[1] =        2	(count) |
  MNS[2] =        2	(count) |
  DTD[0] =      100	(dBar) |
  DTD[1] =       40	(dBar) |
  DTD[2] =       30	(dBar) |
     MDS =     1000	(dBar) |
     APT =        1	(0=dis 1=en) |
     BTM =        1	(Bluetooth Mode) |
     DDB =        0	(0=execute, 1 =bypass) |
     XAS =        2	(1=diff 2=curv) |
     CPS =        1	(0=dis, 1=e) |
     CST =        0	(0=binavg 1=raw) |
     CTF =        1	(dBar) |
     CTL =        0	(dBar) |
     CTU =        2	(dBar) |
     CTC =        0	(C Temp mode) |
     CSP =     1000	(ms) |
  CPB[1] =     2000	(dBar) |
  CPB[2] =     1000	(dBar) |
  CPB[3] =      300	(dBar) |
  CPN[1] =        2	(dBar) |
  CPN[2] =        2	(dBar) |
  CPN[3] =        1	(dBar) |
  CPP[1] =        2	(ms) |
  CPP[2] =        2	(ms) |
  CPP[3] =        1	(ms) |
     CRD =       30	(dBar) |
     CRS =        0	(HRD samples) |
     MTD =       10	(dBar) |
     MTT =   300000	(ms) |
     MTS =  7776000	(sec) |
     MAM =        0	(0=dis 1=s 2=d 3=p 4=a) |
    MAT1 =        1	(sec) |
    MAT2 =      300	(sec) |
     MAP =      600	(sec) |
     ACA =        2	(gain) |
     ACG =      250	(scale) |
     ACR =        4	(sampling rate) |
     PNB =      900	(V) |
     PNC =      300	(inHg) |
     PXC =      700	(inHg) |
     PBL =    43668	(12V cal) |
     PBH =    50958	(14V cal) |
     PP0 =        0	(oil pump 0 mA cal) |
     PP1 =    25960	(oil pump 1000 mA cal) |
     PVL =     1790	(V12 12V cal) |
     PVH =     2089	(V12 14V cal) |
     PV0 =    61614	(vac 0 inHg cal) |
     PV5 =    32000	(vac 5 inHg cal) |
     PC0 =        0	(V12_sig 0mA cal) |
     PC1 =    16600	(V12_sig 1000mA cal) |
     ICM =        0	(0=dis 1=minT 2=avgT 3=medianT) |
     IMM =        0	(ice month mask) |
  ICA[0] =        3	(summer attempts) |
  ICA[1] =        3	(winter attempts) |
  ICB[0] =        0	(summer cycles) |
  ICB[1] =        0	(winter cycles) |
  IMT[0] =       50	(summer message threshold) |
  IMT[1] =       50	(winter message threshold) |
  ILT[0] =      180	(summer time threshold) |
  ILT[1] =      180	(winter time threshold) |
     ICN =       10	(shallow limit for test dBar) |
     ICX =       30	(deep limit for test dBar) |
     ICT =    -1.40	(deg C) |
     ICS =       10	(sec) |
  ISS[0] =        2	(summer 0=disable 1=always 2=urgent) |
  ISS[1] =        2	(winter 0=disable 1=always 2=urgent) |
     ISM =        1	(bit0=depth bit1=GPS bit2=salinity) |
     IST =     0.20	(PSU) |
     ISO =      300	(sec) |
     ISP =        0	(0=disable 1=enbable) |
  IUE[0] =        0	(summer 0=xmitAll 1=threshold) |
  IUE[1] =        0	(winter 0=xmitAll 1=threshold) |
  ICD[0] =      100	(summer descent depth dBar) |
  ICD[1] =      100	(winter descent depth dBar) |
  ICR[0] =       10	(summer minutes) |
  ICR[1] =       10	(winter minutes) |
  ICG[0] =        0	(summer 0=dis 1=enb) |
  ICG[1] =        0	(winter 0=dis 1=enb) |
     ISA =        1	(attempts) |
     ISD =       10	(dBar) |
     IAT =     1200	(secs) |
     ISV =       30	(secs) |
     ISW =     1200	(secs) |
  ICO[0] =      180	(summer secs) |
  ICO[1] =      180	(winter secs) |
     ICF =        0	(summer mask) |
     ICW =        0	(winter ice mask) |
     DNS =      189	(cycle number) |
     DRA =       20	(target descent rate dBar/sec) |
  DSD[0] =       10	(dBar) |
  DSD[1] =       10	(dBar) |
  DSD[2] =        8	(dBar) |
  DST[0] =     1800	(sec) |
  DST[1] =     1800	(sec) |
  DST[2] =     1800	(sec) |
     DDF =      300	(sec) |
     DDC =        0	(sec) |
     DSI =     1000	(ms) |
     DDD =       10	(dBar) |
     DDT =     3600	(sec) |
  DFT[0] =      300	(sec) |
  DFT[1] =      310	(sec) |
  DFT[2] =      385	(sec) |
     DCM =        1	(0=dis 1=en) |
     DAL =        1	(0=cal 1=seek) |
  DCT[0] =      900	(sec) |
  DCT[1] =     1800	(sec) |
  DCT[2] =     1800	(sec) |
     DCA =        3	(max attempts) |
     DRC =        3	(cal rate calculation) |
     DRT =        0	(sec) |
     DCX =     1020	(sec) |
     DCP =        1	(sec) |
     DCF =        1	(sec) |
     DCD =      200	(dBar) |
     DCR =        0	(dBar/sec) |
     DSW =     7200	(sec) |
     DUC =      100	(under damping factor) |
     DPT =       10	(sec) |
     DUT =       10	(sec) |
     DPD =        5	(sec) |
     DVT =       30	(sec) |
     DVD =        5	(dBar) |
     DVW =     1800	(sec) |
     PFR =       19	(cc/min) |
     PFC =       16	(.001 cc/dBar) |
     DDP =       50	(dBar) |
     DDS =     1800	(sec) |
     PNS =      189	(cycle) |
  PTO[0] =        0	(sec) |
  PTO[1] =     7200	(sec) |
  PTO[2] =        0	(sec) |
  PTH[0] =        0	(24 LSB) |
  PTH[1] =        0	(24 LSB) |
  PTH[2] =  2097152	(24 LSB) |
     PTD =        0	(sec) |
     PSA =        1	(0=avg 1=avg+raw) |
     PSI =   600000	(ms) |
  PFM[0] =        0	(0=dis 1=en) |
  PFM[1] =        0	(0=dis 1=en) |
  PFM[2] =        1	(0=dis 1=en) |
  PFT[0] =        1	(sec) |
  PFT[1] =      300	(sec) |
  PFT[2] =      300	(sec) |
  PFD[0] =      100	(dBar) |
  PFD[1] =      100	(dBar) |
  PFD[2] =       40	(dBar) |
  PFV[0] =        1	(sec) |
  PFV[1] =        1	(sec) |
  PFV[2] =        5	(sec) |
     PDT =     1800	(sec) |
     PSM =        0	(0=dis 1=en) |
     PSD =      200	(dBar) |
     PST =    10800	(sec) |
     PSV =       30	(sec) |
     PSW =      600	(sec) |
     ASN =      189	(cycle number) |
     ARR =        9	(target rise rate cm/sec) |
     APM =     2400	(sec) |
     APN =       20	(sec) |
     APF =       40	(sec) |
     ASI =     1000	(ms) |
     ASD =        1	(dBar) |
     ADD =       10	(dBar) |
     ADT =     3600	(sec) |
     ARC =     4500	(?) |
     AHR =     1000	(cm/sec) |
     AHD =       10	(dBar) |
     SFN =      189	(cycle number) |
     SDC =        0	(dBar) |
     SGT =       15	(min) |
     SET =       15	(min) |
     SBM =        0	(0=dis 1=en) |
     SBT =      300	(sec) |
     SDT =        0	(interval) |
     SSA =        0	(0=dis 1=en) |
     SSI =     1000	(ms) |
     SDS =       10	(sec) |
     SAM =        1	(0=dis 1=en) |
     SAX =     7500	(.001 psi) |
     SAN =     6500	(.001 psi) |
     SAO =     1080	(sec) |
     SAP =     1200	(sec) |
     SPT =      300	(sec) |
     SSP =      240	(sec) |

