Type: 1
P ref: 112    1
T   ( 1 )  25861  20.861 
T   ( 2 )  24851  19.851 
T   ( 3 )  24530  19.530 
T   ( 4 )  24297  19.297 
T   ( 5 )  24067  19.067 
T   ( 6 )  23784  18.784 
T   ( 7 )  23493  18.493 
Type: 0
P ref: 112    1
S   ( 1 )  32921  31.921 
S   ( 2 )  32896  31.896 
S   ( 3 )  32881  31.881 
S   ( 4 )  32872  31.872 
S   ( 5 )  32862  31.862 
S   ( 6 )  32849  31.849 
S   ( 7 )  32825  31.825 
Type: 21
P   ( 1 )  112     1.2 
Type: 5
P:     8.9 
T:  16.724 
S:  31.799 
P deviation: -383 
Type: 4
Lat:   41.5342      41.5345
Lon:  -70.6468     -70.6466
Flt/GPS day:    6803      6804         0         0   2014/05/25   2014/05/26        -            -    
Flt/GPS time:  63096      4959     63097      4959     17:31:36     01:22:39     17:31:37     01:22:39
Loop count: 8    5
P drift:  9.0     8.9
T drift: 16.705    16.723
S drift: 31.793    31.800
First bin P: 169
First bin T: 23493
First bin S: 32825
Last bin P: 112
Number of bins: 7
New offset P: 0
Step 1 P: 37
Step 2 P: 90
Total pumping time out/in: 0 0    3069 0
Auxillary Battery voltage in dV: 152    146    147
CPU Battery voltage in dV: 104    103    103
Internal Pressure is: 79    79    68
System flag: 1  81  3
Profile status: 2a  48  1d
Day for engineering packets:    6804   6804   6804   2014/05/26   2014/05/26   2014/05/26
Time for engineering packets:   3000   4105   4705     00:50:00     01:08:25     01:18:25
Surface tries: 1
Modem Faults: 0
SBD Failures: 0
Cycle Number: 2
