Type: 1
P ref: 112    1
T   ( 1 )  25891  20.891 
T   ( 2 )  25889  20.889 
T   ( 3 )  25891  20.891 
T   ( 4 )  25892  20.892 
T   ( 5 )  25880  20.880 
T   ( 6 )  25848  20.848 
T   ( 7 )  25793  20.793 
Type: 0
P ref: 112    1
S   ( 1 )  32979  31.979 
S   ( 2 )  32979  31.979 
S   ( 3 )  32978  31.978 
S   ( 4 )  32977  31.977 
S   ( 5 )  32975  31.975 
S   ( 6 )  32972  31.972 
S   ( 7 )  32972  31.972 
Type: 21
P   ( 1 )  112     1.2 
Type: 5
P:     8.8 
T:  20.591 
S:  31.962 
P deviation: -383 
Type: 4
Lat:   41.5340      41.5342
Lon:  -70.6467     -70.6466
Flt/GPS day:    6940      6941         0         0   2014/10/09   2014/10/10        -            -    
Flt/GPS time:  68714      4086     68714      4087     19:05:14     01:08:06     19:05:14     01:08:07
Loop count: 4    11
P drift:  8.7     8.8
T drift: 20.583    20.591
S drift: 31.990    31.962
First bin P: 168
First bin T: 25793
First bin S: 32972
Last bin P: 112
Number of bins: 7
New offset P: 0
Step 1 P: 71
Step 2 P: 9
Total pumping time out/in: 0 0    280 288
Auxillary Battery voltage in dV: 153    150    150
CPU Battery voltage in dV: 104    103    103
Internal Pressure is: 81    81    70
System flag: 1  81  3
Profile status: 2a  4a  15
Day for engineering packets:    6941   6941   6941   2014/10/10   2014/10/10   2014/10/10
Time for engineering packets:   2100   2367   3642     00:35:00     00:39:27     01:00:42
Surface tries: 1
Modem Faults: 1
SBD Failures: 0
Cycle Number: 35
