Type: 1
P ref: 111    1
T   ( 1 )  25875  20.875 
T   ( 2 )  25876  20.876 
T   ( 3 )  25876  20.876 
T   ( 4 )  25875  20.875 
T   ( 5 )  25873  20.873 
T   ( 6 )  25854  20.854 
T   ( 7 )  25787  20.787 
Type: 0
P ref: 111    1
S   ( 1 )  32974  31.974 
S   ( 2 )  32974  31.974 
S   ( 3 )  32974  31.974 
S   ( 4 )  32974  31.974 
S   ( 5 )  32973  31.973 
S   ( 6 )  32970  31.970 
S   ( 7 )  32964  31.964 
Type: 21
P   ( 1 )  111     1.1 
Type: 5
P:     8.8 
T:  20.569 
S:  31.961 
P deviation: -383 
Type: 4
Lat:   41.5342      41.5341
Lon:  -70.6472     -70.6468
Flt/GPS day:    6940      6940         0         0   2014/10/09   2014/10/09        -            -    
Flt/GPS time:  25641     46686     25642     46686     07:07:21     12:58:06     07:07:22     12:58:06
Loop count: 11    11
P drift:  8.7     8.8
T drift: 20.590    20.568
S drift: 31.977    31.961
First bin P: 168
First bin T: 25787
First bin S: 32964
Last bin P: 111
Number of bins: 7
New offset P: 0
Step 1 P: 79
Step 2 P: 3
Total pumping time out/in: 0 0    280 289
Auxillary Battery voltage in dV: 153    150    150
CPU Battery voltage in dV: 104    103    103
Internal Pressure is: 81    81    70
System flag: 1  81  3
Profile status: 2a  4a  15
Day for engineering packets:    6940   6940   6940   2014/10/09   2014/10/09   2014/10/09
Time for engineering packets:  44700  44967  46241     12:25:00     12:29:27     12:50:41
Surface tries: 1
Modem Faults: 1
SBD Failures: 0
Cycle Number: 35
