Type: 1
P ref: 112    1
T   ( 1 )  29093  24.093 
T   ( 2 )  29031  24.031 
T   ( 3 )  28997  23.997 
T   ( 4 )  28978  23.978 
T   ( 5 )  28960  23.960 
T   ( 6 )  28937  23.937 
T   ( 7 )  28910  23.910 
Type: 0
P ref: 112    1
S   ( 1 )  32584  31.584 
S   ( 2 )  32584  31.584 
S   ( 3 )  32584  31.584 
S   ( 4 )  32584  31.584 
S   ( 5 )  32583  31.583 
S   ( 6 )  32583  31.583 
S   ( 7 )  32584  31.584 
Type: 21
P   ( 1 )  112     1.2 
Type: 5
P:     8.9 
T:  23.800 
S:  31.577 
P deviation: -383 
Type: 4
Lat:   41.5342      41.5343
Lon:  -70.6468     -70.6468
Flt/GPS day:    6885      6885         0         0   2014/08/15   2014/08/15        -            -    
Flt/GPS time:  47569     68560     47569     68560     13:12:49     19:02:40     13:12:49     19:02:40
Loop count: 7    5
P drift:  8.9     8.9
T drift: 23.794    23.800
S drift: 31.580    31.577
First bin P: 167
First bin T: 28910
First bin S: 32584
Last bin P: 112
Number of bins: 7
New offset P: 0
Step 1 P: 82
Step 2 P: 4
Total pumping time out/in: 0 0    1260 615
Auxillary Battery voltage in dV: 153    150    150
CPU Battery voltage in dV: 104    103    104
Internal Pressure is: 80    80    68
System flag: 1  81  3
Profile status: 2a  48  15
Day for engineering packets:    6885   6885   6885   2014/08/15   2014/08/15   2014/08/15
Time for engineering packets:  66300  67042  68205     18:25:00     18:37:22     18:56:45
Surface tries: 1
Modem Faults: 1
SBD Failures: 0
Cycle Number: 38
