Type: 1
P ref: 113    1
T   ( 1 )  29882  24.882 
T   ( 2 )  29827  24.827 
T   ( 3 )  29666  24.666 
T   ( 4 )  29527  24.527 
T   ( 5 )  29391  24.391 
T   ( 6 )  29252  24.252 
T   ( 7 )  29204  24.204 
T   ( 8 )  29193  24.193 
Type: 0
P ref: 113    1
S   ( 1 )  20872  19.872 
S   ( 2 )  20864  19.864 
S   ( 3 )  20849  19.849 
S   ( 4 )  20846  19.846 
S   ( 5 )  20845  19.845 
S   ( 6 )  20847  19.847 
S   ( 7 )  20848  19.848 
S   ( 8 )  0  -1.000 
Type: 21
P   ( 1 )  113     1.3 
Type: 5
P:     8.7     8.7 
T:  24.420  24.410 
S:  19.848  19.848 
P deviation: -54 -54 
Type: 4
Lat:   41.5343      41.5344
Lon:  -70.6467     -70.6468
Flt/GPS day:    5815      5816      5815      5816   2011/09/10   2011/09/11   2011/09/10   2011/09/11
Flt/GPS time:  59054     56497     59054     56497     16:24:14     15:41:37     16:24:14     15:41:37
Loop count: 2    6
P drift:  8.7     8.7
T drift: 24.463    24.204
S drift: 19.850    19.849
First bin P: 177
First bin T: 29193
First bin S: 20848
Last bin P: 113
Number of bins: 8
New offset P: 0
Step 1 P: 10
Step 2 P: 68
Total pumping time out/in: 0 0    3250 0
Auxillary Battery voltage in dV: 154    148    149
CPU Battery voltage in dV: 105    103    105
Internal Pressure is: 81    81    70
System flag: 1  81  3
Profile status: 2a  48  1d
Day for engineering packets:    5816   5816   5816   2011/09/11   2011/09/11   2011/09/11
Time for engineering packets:  54900  55621  56221     15:15:00     15:27:01     15:37:01
Surface tries: 1
Modem Faults: 0
SBD Failures: 8
Cycle Number: 3
